- Methodology article
- Open Access
Real-time determination of intracellular oxygen in bacteria using a genetically encoded FRET-based biosensor
BMC Biology volume 10, Article number: 28 (2012)
Molecular oxygen (O2) is one of the key metabolites of all obligate and facultative aerobic pro- and eukaryotes. It plays a fundamental role in energy homeostasis whereas oxygen deprivation, in turn, broadly affects various physiological and pathophysiological processes. Therefore, real-time monitoring of cellular oxygen levels is basically a prerequisite for the analysis of hypoxia-induced processes in living cells and tissues.
We developed a genetically encoded Förster resonance energy transfer (FRET)-based biosensor allowing the observation of changing molecular oxygen concentrations inside living cells. This biosensor named FluBO (fluorescent protein-based biosensor for oxygen) consists of the yellow fluorescent protein (YFP) that is sensitive towards oxygen depletion and the hypoxia-tolerant flavin-binding fluorescent protein (FbFP). Since O2 is essential for the formation of the YFP chromophore, efficient FRET from the FbFP donor domain to the YFP acceptor domain only occurs in the presence but not in the absence of oxygen. The oxygen biosensor was used for continuous real-time monitoring of temporal changes of O2 levels in the cytoplasm of Escherichia coli cells during batch cultivation.
FluBO represents a unique FRET-based oxygen biosensor which allows the non-invasive ratiometric readout of cellular oxygen. Thus, FluBO can serve as a novel and powerful probe for investigating the occurrence of hypoxia and its effects on a variety of (patho)physiological processes in living cells.
Non-invasive detection of intracellular O2 is of particular importance since it is one of the key metabolites of obligate and facultative aerobic organisms. Cellular O2 is a prominent indicator for oxygen-dependent metabolic activities, such as aerobic respiration or oxygen dependent synthesis and degradation of cellular components [1, 2]. In addition, various biological, pathological and biotechnological processes are controlled by O2 limitation, including biofilm formation and host-pathogen interactions [3–7], hypoxia induced inflammatory processes , tumor pathophysiology [9–12] as well as microbial fermentation processes used for bioremediation and the production of food, feed and biofuels [13–16].
To date, different minimally invasive fluorescence and phosphorescence based O2-sensitive probes have been developed for imaging molecular oxygen in cells and tissues. Among them, platinum(II)-porphyrin dyes are widely used for analyzing hypoxia-induced responses of mammalian cells [17–19]. Alternatively, the green fluorescent protein (GFP) and its variants can be applied as genetically encoded intracellular probes that are specifically expressed and can be selectively targeted within defined cells and tissues. In this context, at least two 'passive' GFP-based oxygen sensors have been developed for estimating intracellular oxygen levels in E. coli. Here, GFP was applied as a reporter protein expressed under control of oxygen-responsive E. coli promoters [20, 21]. Additionally, oxygen sensitive photoactivation of GFP-mediated red fluorescence was applied for in vivo imaging of oxygen in mammalian cells and organs [22–24].
Remarkably, molecular oxygen can currently not be analyzed in vivo by genetically encoded FRET-based biosensors, although these biosensors constitute one of the most widespread classes of fluorescent molecular probes used for the non-invasive quantitative analysis of intracellular compounds including Ca2+, Zn2+, Cl-, pH, H2O2, ATP, maltose, sucrose, ribose and glucose [25–27]. With respect to oxygen sensing, however, GFP-like proteins, which are commonly used as donor and acceptor domains of FRET biosensors, exhibit a major drawback: Their autocatalytic chromophore synthesis strictly depends on the presence of molecular oxygen [28, 29] and thus the fluorescence signal intensity basically does not reflect the amount of synthesized reporter protein . Therefore, GFP and its color variants can not solely be used as fluorescent biosensor domains for accurate O2 determination.
Recently, we developed a novel class of fluorescent proteins which carry flavin mononucleotide (FMN) as chromophore . In contrast to GFP-like FPs, the fluorescence signal of these FMN-based fluorescent proteins (FbFP) is independent of cellular oxygen and thus FbFP can be used as quantitative in vivo real-time reporter protein under aerobic as well as anaerobic conditions [30, 31]. Here, we report the construction and application of the first genetically encoded FRET-based biosensor for oxygen named FluBO which consists of the oxygen-insensitive FbFP donor domain and the hypoxia-sensitive enhanced yellow fluorescent protein (YFP) acceptor domain. We further show that its FRET efficiency dynamically responds to changing O2 values in living bacterial cells.
Construction and in vitrocharacterization of FluBO
The in vivo fluorescence of YFP, in contrast to FbFP, strictly depends on intracellular oxygen . This observation prompted us to test if a YFP-FbFP fusion can be used as a ratiometric FRET-based biosensor for self-referenced determination of molecular oxygen levels in living cells. The canonical intramolecular FRET-based biosensors are generally fusion proteins consisting of two fluorescent domains with different chromophores that show spectral overlap between the absorption wavelengths of the acceptor domain and the emission wavelengths of the donor domain [32, 33]. Therefore, we constructed a recombinant gene (Figure 1A, Additional file 1) encoding the oxygen biosensor which consists of the N-terminal YFP acceptor domain and the C-terminal FbFP donor domain interconnected by a small peptide linker with a thrombin protease cleavage site. Since the enhanced cyan FP (CFP) has most widely been used together with YFP in a tandem FRET biosensor design [32, 34], we initially tested whether CFP can be functionally substituted by FbFP as the FRET donor domain. For that purpose, YFP and FbFP as well as the YFP-FbFP fusion protein were separately expressed, purified and their spectral characteristics were analyzed. Figure 1B shows the absorption and emission spectra of FbFP and YFP, respectively. FbFP exhibited its typical absorption spectrum ranging from near UV to blue light with λmax at 450 nm, which is characteristic for the FMN chromophore . As required for suitable FRET pairs, the fluorescence emission spectrum of FbFP (λmax = 495 nm) substantially overlaps with the YFP absorption spectrum (λmax = 512 nm) whereas the emission peaks of FbFP and YFP are sufficiently separated (YFP λmax = 528 nm). To minimize direct excitation of the acceptor at the donor excitation wavelength, two-dimensional wavelength scans of FbFP and YFP have been carried out. As demonstrated in Figure 1C, an excitation wavelength of 380 nm results in a bright FbFP fluorescence and a very low excitation of YFP fluorescence. Beside its spectral characteristics, the quantum yield (QY) of FbFP  (QY = 0.39) resembles that of CFP  (QY = 0.36) indicating that FbFP may be suitable as a FRET donor for YFP.
FRET is a phenomenon of nonradiative energy transfer between the chromophores of the donor and acceptor domain, which can be observed either by a change of (i) ratio of fluorescence intensities emitted by each domain or (ii) lifetime of donor fluorescence. Efficient intramolecular FRET thus results in a decrease of donor fluorescence intensity and of excited-state lifetime. To demonstrate that the YFP-FbFP fusion protein forms a functional FRET pair we first analyzed its fluorescence properties in vitro. Fluorescence emission spectra (Figure 1D) of FbFP, YFP and FluBO, respectively, recorded with an excitation wavelength of 380 nm clearly demonstrate that the FbFP fluorescence (green) is largely decreased by efficient energy transfer leading to a dominant fluorescence of the YFP acceptor domain within the fusion protein (red). In contrast, fluorescence intensity of the plain acceptor was low at the same excitation wavelength (yellow). Protease cleavage of the two fluorescent domains resulted in distances much larger than the Förster radius between the donor and acceptor domain and thus in the loss of FRET coupling, which is reflected by a dequenching of the donor FbFP and a dramatic decrease of YFP fluorescence (see Additional file 2).
In vivofunctionality of FluBO
Next, FluBO was tested as a FRET-based ratiometric oxygen probe in living cells. Initially, we analyzed the FbFP emission of FluBO in E. coli by fluorescence lifetime imaging (FLIM) under aerobic conditions leading to complete maturation of the YFP (that is, the FluBO acceptor domain) chromophore. Figure 2A shows E. coli cells expressing either FbFP or FluBO color-coded according to their average fluorescence lifetimes observed at the FbFP fluorescence maximum. Reduction of in vivo fluorescence lifetimes of plain FbFP (monoexponential analysis, τave = 2.73 ns) in comparison to FbFP fused to the FRET acceptor domain YFP (biexponential analysis, τave = 1.74 ns) (Figure 2A, B) again revealed efficient energy transfer between the donor and the acceptor domain of FluBO with a difference in fluorescence lifetime (Δτ) of 0.99 ns under aerobic conditions. The apparent FRET efficiency (E) is 37% using τave (Figure 2B). A detailed analysis of the FluBO-FLIM experiments in aerated E. coli cells revealed that two exponential functions are necessary to describe the observed fluorescence decay behavior satisfactorily (Figure 2C). Here, the longer fluorescence lifetime is very similar to that of FbFP alone. The other, shorter lifetime (mean τ1 = 1.03 ns) exhibiting the major amplitude (mean relative amplitude of 67%) represents the FRET process (donor to acceptor energy transfer). Hence, a FRET efficiency of approximately 62% within the FluBO proteins can be deduced from τ1. In comparison, the original FRET donor CFP exhibits a similar fluorescence lifetime with a τave of 2.23 ns whereas FRET-mediated quenching led to an average fluorescence lifetime of 1.67 ns within a YFP-CFP fusion (Δτ = 0.56 ns) and an apparent efficiency E of 25.0% .
Afterwards, we studied the oxygen-dependent fluorescence response of FluBO in E. coli cells during batch cultivation (Figure 3A). To this end, the BioLector microbioreactor system (m2p-labs, Aachen, Germany) was used, which provides quantitative online data of the cell density via scattered light intensity (black curve) and dissolved oxygen tension (DOT; blue curve) as described previously [30, 36, 37]. In parallel, intracellular oxygen values of E. coli cells expressing FluBO were permanently recorded during cultivation by measuring the donor fluorescence at 492 nm (FbFP, green) and the acceptor fluorescence at 532 nm (YFP, yellow) with an excitation wavelength of 380 nm. After approximately three hours of cultivation, FluBO expression was autoinduced as indicated by rapidly increasing fluorescence intensities of both FP domains. After five hours (to), high respiratory activity of exponentially growing bacteria caused a drastic limitation of O2 in the growth medium (DOT ≤ 5%). During this phase of growth, concomitant intracellular oxygen depletion was reflected by a weak increase of YFP fluorescence due to permanently increasing cell numbers and accumulated biosensor protein at low O2 concentrations. After 13 hours of cell cultivation, the change of carbon source from glycerol to the overflow metabolite acetate, produced in the first phase of the fermentation, led to a reduced oxygen consumption of E. coli cells. Thus, the diauxic growth temporarily elevated the intracellular oxygen value (Figure 3A, t1-t2) which could be monitored instantly by an increasing YFP fluorescence accompanied by FRET-based decrease of FbFP fluorescence. Consequently, reduced cell respiration led to a transient shift of DOT to a maximal level of 11%. Finally, the DOT level rapidly increased after 16.0 hours when the culture entered the stationary growth phase (t3). As expected, this final change of the oxygen concentration also caused an immediate increase of YFP fluorescence accompanied by a decrease of FbFP fluorescence.
The YFP-FbFP fusion construct FluBO represents a FRET pair and thus provides a fixed stoichiometry of the two FP domains. Hence, the ratio of observed YFP and FbFP fluorescence emissions directly reflects the change of cellular oxygen concentration (Figure 3B). Remarkably, extra- and intracellular O2 values are directly correlated and developed cooperatively during cell cultivation. For instance, low DOT which occurred in the late logarithmic growth phase (Figure 3A, B; t2-t3) resulted in an almost constant YFP-to-FbFP fluorescence ratio (Figure 3B) until the molecular oxygen concentration increased due to the physiological adaptation of the bacteria during the stationary growth phase. Here, the change of intracellular oxygen availability is directly reflected by an increasing FluBO fluorescence ratio. The ratiometric change of donor and acceptor fluorescence is caused by O2-mediated maturation of the YFP chromophore resulting in dominant YFP fluorescence and a FRET-dependent quenching of FbFP fluorescence (Figure 3A, B). However, bacterial proteins exhibiting an intrinsic fluorescence may affect the biosensor-based oxygen readout. Therefore, an E. coli strain carrying the corresponding empty expression vector was cultivated under the same conditions and the ratio of cell-mediated background fluorescence was analyzed at identical wavelengths (Figure 3C). Here, an only marginal change of the emission ratio was detected, which was clearly independent of the DOT level.
In vivocalibration of the fluorescent protein-based oxygen biosensor
The results presented so far clearly demonstrate that the development of DOT directly correlates with intracellular oxygen levels basically allowing the in vivo calibration of the FRET biosensor within the used E. coli test system. However, the emission ratio of FluBO at a certain time point of cell growth is modulated by the availability of oxygen as well as by the time needed for FluBO synthesis and YFP chromophore formation. Furthermore, oxygen-dependent maturation of the YFP chromophore is an irreversible process. Therefore, repeating changes from sufficient to deficient oxygen concentrations as occurring during batch cultivation of E. coli indeed resulted in the inhibition of YFP chromophore maturation and thus led to almost constant but stepwise increased cyan-to-yellow fluorescence ratios under anaerobic conditions (Figure 3B). For that reason, we subsequently considered the change of FluBO fluorescence ratio over time (d (ratio)/d (t)) as a measure for fluctuating intracellular O2 concentration. To accurately calibrate FluBO in vivo, E. coli cells were batch cultured to the mid-logarithmic growth phase (that is, to oxygen limited conditions) and, subsequently, DOT was temporarily increased twice. As shown in Figure 4A the change of FluBO emission ratio over time directly corresponded to the respective DOT levels. Remarkably, with the onset of repeated O2 deprivation in the medium, lowering of intracellular oxygen level led to a retarded YFP chromophore maturation, which in turn was directly reflected by a decreasing (d (ratio)/d (t)) value of FluBO fluorescence. Hence, this new method also allows the indirect detection of decreasing oxygen levels in living cells. Finally, a linear oxygen calibration curve could be derived from the E. coli in vivo fluorescence data (Figure 4B) covering a DOT level from 0 to 33% corresponding to a molecular O2 concentration up to 0.08 mmol/l.
Characteristics of the oxygen biosensor FluBO
Here, we describe for the first time the development of a genetically encoded FRET-based biosensor, allowing the determination of molecular oxygen inside live cells. The biosensor FluBO consists of the cyan fluorescing flavoprotein FbFP and the yellow fluorescing GFP derivative YFP as respective FRET donor and acceptor domains. Here, the oxygen demand for chromophore formation of YFP, but not of FbFP is the basis for the detection of cellular oxygen changes via the O2-dependent alteration of FRET efficiency. At atmospheric levels of O2, YFP chromophore maturation leads to efficient FRET coupling causing an efficient quenching of FbFP fluorescence and a dominant YFP fluorescence. In contrast, the absence of molecular oxygen prevents YFP chromophore formation and thus immature YFP has no effect on the fluorescence of the FbFP donor domain due to the loss of FRET coupling. To evaluate the in vivo functionality of FluBO, we extensively analyzed the biosensor in E. coli cells during batch cultivation. Parallel online detection of extra- and intracellular oxygen levels using an oxygen sensitive optode and the genetically encoded biosensor revealed a clear correlation between O2 levels outside and inside the cells.
Calibration of FluBO
Development of DOT directly correlates with intracellular O2 levels hence allowing the in vivo calibration of the FRET biosensor within the used E. coli test system. The resulting oxygen calibration curve was derived from the FluBO (d (ratio)/d (t)) values that correspond to defined concentrations of extracellular oxygen.
Generally, novel intramolecular FRET-based biosensors are pre-calibrated in vitro and subsequently transferred into the living system of interest [38–41]. However, in contrast to other FP-based biosensors which allow the quantification of a certain metabolite, in vitro calibration of FluBO is not an easy task. Accurate pre-calibration requires biosensor protein, which was expressed and subsequently purified under anaerobic conditions. In addition, in vitro calibration has to be conducted using the gaseous substrate O2 in an otherwise oxygen-free atmosphere. Furthermore, biosensor pre-calibration basically does not take into account that differences in non-specific interactions between a molecule (that is, the respective biosensor) and its immediate surrounding (for example, a buffer in comparison to the interior of a living cell) can greatly influence the equilibrium and rate of the respective reaction [42, 43]. For that reason, the use of pre-calibrated reference probes does not inevitably lead to accurate in vivo calibration of a novel biosensor. Finally, as discussed below, the level of biosensor expression and, to a minor extent, its stability (or turnover) in the respective host cell rather than the cellular growth rate most probably represents the major bottleneck for functionality of a genetically encoded biosensor. However, these factors significantly depend on the individual properties of a biological system and might even vary within a particular cell depending on growth conditions and/or environmental parameters. Additionally, ratio of extra- and intracellular oxygen concentrations in different organisms might vary in dependence on O2 consumption, which in turn can also be affected by individual parameters, such as variable metabolic activities and cell organization as well as heterogeneity of oxygen diffusion within cellular compartments and tissues. Consequently, the intracellular O2 concentration inside E. coli cells is probably much lower than in their surrounding medium (reflected by DOT). Therefore, the applicability of an intracellular biosensor as a quantitative probe needs to be evaluated and calibrated separately for each organism and condition to achieve reliable data.
Environmental sensitivity of FluBO
Most of the YFP derivatives described in the literature exhibit a distinct pH sensitivity with pKa values ranging from 6.9 (EYFP) to 5.7 (mCitrine) . Using YFP as a FRET receptor domain, therefore, ideally requires a constant fluorescence brightness which is not affected by intracellular changes of pH. To quantitatively interpret oxygen measurements with FluBO, we utilized an EYFP variant distributed by Clontech-Takara (details are described within the Methods section and the corresponding EYFP sequence is shown in Additional file 1). Analysis of pH sensitivity demonstrated that this YFP derivative exhibits a remarkable pH resistance (pKa = 5.2; Additional file 3). Thus, FluBO-based oxygen determination is not affected within the physiological pH range. In addition, the early YFP variants were also sensitive towards chloride . To further rule out that chloride ions influenced the FluBO oxygen response in vivo, we also determined Cl- sensitivity of the used EYFP. We could demonstrate that YFP fluorescence intensity remained constant in the presence of chloride concentrations up to 100 mM (data not shown). This result clearly indicates that FluBO is also insensitive to changes of chloride-ion concentrations in the physiological relevant range.
Limitations of FluBO
Although FluBO is a useful molecular probe, some aspects must be considered to allow online-detection of intracellular oxygen: First, as mentioned above, oxygen-dependent chromophore formation is an irreversible process. Consequently, this biosensor indeed allows analyzing changes from low to high oxygen concentrations but it cannot directly be used to detect reduction of oxygen levels via its change of intramolecular FRET efficiency. Therefore, we analyzed alterations from sufficient to deficient oxygen levels by observing the total changes of dual emission ratio over time. By using this method, recurring oxygen limitation was reflected by a stagnating alteration of YFP-to-FbFP fluorescence ratio (as for example, visible in Figure 3B between t2 and t3) and in turn by decreasing (d (ratio)/d (t)) values (Figure 4A). Furthermore, it should be emphasized that due to the irreversibility of O2-mediated YFP chromophore formation repeated changes or a long-lasting increase of intracellular oxygen level can be determined only under specific conditions. Namely, the degradation of oxidized FluBO and the production rate of non-oxidized FluBO need to form an equilibrium ensuring a continuous and sufficient supply of biosensor molecules which did not 'sense' oxygen before. It, therefore, remains to be experimentally demonstrated how FluBO behaves in cells exhibiting significantly lower metabolic activity than E. coli. Nevertheless, we expect the FluBO (d (ratio)/d (t)) value to represent a valid indication of intracellular oxygen levels in a large variety of different (micro)organisms. To this end, we are currently analyzing the applicability of FbFP and the FbFP-based biosensor FluBO in various prokaryotic and eukaryotic including mammalian cell lines.
Alternative in vivo O2sensors
Monitoring the heterogeneity and development of intracellular oxygen levels with high spatio-temporal resolution requires a molecular O2 biosensor that exhibits some important properties. For example, the ideal oxygen indicator must be either uniformly distributed or, alternatively, can be specifically targeted within a cell or tissue of interest. For quantitative in vivo O2 determination, the biosensor has to respond rapidly, reliably and reversibly to the full range of physiological oxygen concentrations.
In the last decade, different fluorescent dyes and proteins have been developed, which can be used as sensors for intracellular oxygen measurements.
(1) Various fluorescence and phosphorescence based O2-sensitive probes, including the complex platinum(II)-porphyrin dyes are widely used for imaging molecular oxygen in living organisms [17–19]. These probes exhibit a selective and reversible response to O2 within the full physiological oxygen range (0 to 250 μM) combined with optimal photophysical properties (for example, high fluorescence brightness). Nevertheless, in comparison to genetically encoded sensors intracellular loading of these dyes appears to be relatively inefficient, slow and, furthermore, they cannot be generally targeted to specific cells or intracellular compartments [19, 44, 45]. Consequently, intracellular visualization of O2 with high spatio-temporal resolution is limited.
(2) Alternatively, GFP can be utilized as a reporter gene, whose expression is under control of an oxygen sensitive regulator within a certain organism. For example, in E. coli promoters P nar (controlling nitrate reductase expression) and P rpoS (controlling the expression of the stress response sigma factor σS), which are activated under microaerobic and anaerobic conditions, were applied to monitor oxygen limitations [20, 21]. Although this technique is potentially capable of analyzing limited oxygen supply and differences in population heterogeneities, this biosensor architecture can only be applied for specific host organisms. Furthermore, regulators are often part of a higher-order regulatory network and thus might not solely respond to oxygen deprivation. Beside oxygen sensitive GFP reporter gene constructs, Takahashi and coworkers described an exceptional technique that also allows GFP-dependent in vivo oxygen determination [22–24]. This approach makes use of the oxygen sensitive red-shifted fluorescence of GFP that occurs after photoactivation with blue light . However, since the magnitude of the observed red shift is abruptly increased with oxygen concentrations < 1%, this technique can only be applied for measuring low intracellular oxygen levels. Furthermore, the basic mechanism of GFP photoactivation of red fluorescence as well as the oxygen-dependent recovery to green fluorescence is not fully understood yet. Finally, it is worth mentioning that both of the described strategies, using GFP either as 'passive' or 'active' oxygen biosensor, inherently suffer from impaired chromophore formation under oxygen limited conditions. This fact makes FluBO the first fluorescent protein-based biosensor enabling exact and reliably determination of molecular oxygen.
Accurate determination of intracellular oxygen and analysis of changing O2 levels inside living cells and tissues is of the utmost importance in gaining new insights into complex (patho)physiological processes associated with oxygen deprivation, including invasion and persistence of pathogens or tumor development. Our results show that FluBO can be used as a ratiometric biosensor for molecular oxygen that changes the emission maxima of the donor and acceptor domain as a function of O2 availability and its sensitivity is within the physiological range relevant for bacteria as well as for eukaryotes, including mammalian and human cells . The oxygen modulated FRET-coupling allows the non-invasive readout of cellular O2 levels as well as the online analysis of its changes with high spatial and temporal resolution by fluorescence spectrometry and FLIM. Thus, the observation of cellular oxygen levels in any cellular compartment using genetically encoded FRET biosensors will offer a novel rational approach to understand hypoxia-dependent processes in biological, biomedical and biotechnological processes.
Bacteria and growth conditions
E. coli strain DH5α  was used for DNA cloning of the expression vectors encoding the fluorescent reporter proteins FbFP and YFP, as well as the biosensor FluBO. E. coli strain BL21(DE3) (Novagen, distributed by Merck KGaA, Darmstadt, Germany) was used for expression of the fluorescent proteins and FRET-based biosensor FluBO. For protein expression and purification, bacterial cells were grown either in autoinduction TB medium consisting of 5 g/l glycerol, 12 g/l tryptone, 24 g/l yeast extract, 2.32 g/l KH2PO4, 12.5 g/l K2HPO4 (pH 7.2), lactose 2 g/l, glucose 0.5 g/l (for FbFP expression and purification) or in LB medium consisting of 10 g/l tryptone, 5 g/l yeast extract, 10 g/l NaCl (for YFP expression and purification). BioLector cultivation experiments were conducted in Overnight Express™ Instant TB medium (Novagen) as described below. All media were supplemented with 50 μg/l kanamycin to maintain expression vector pRhotHi-2.
Construction of FP expression vectors
The construction of expression vector pRhotHi-2-YFP encoding the enhanced yellow fluorescent protein YFP (available from Clontech-Takara Bio Europe, Saint-en-laye, France) and pRhotHi-2-FbFP encoding EcFbFP (GenBank number: ABN71355; commercially available as evoglow-Bs2, evocatal GmbH, Düsseldorf, Germany) was described previously . The YFP variant used in this study contains four amino acid substitutions previously published as GFP-10C , as well as the substitution H231L and the insertion of valine behind the first amino acid. The expression vector pRhotHi-2-FluBO harboring the FRET-based oxygen biosensor encoding gene was constructed using a synthetic FluBO gene (Eurofins MWG Operon, Ebersberg, Germany) whose DNA sequence is shown in Additional file 1. The biosensor gene was cloned into the NdeI and XhoI restriction sites of pRhotHi-2  and recombinant FP genes were confirmed by DNA sequencing. The pRhotHi-2-FluBO vector will soon be commercially available (evocatal GmbH, Düsseldorf, Germany).
Protein expression and purification
E. coli BL21(DE3) cells carrying the expression plasmids pRhotHi-2-FbFP, pRhotHi-2-YFP and pRhotHi-2-FluBO, respectively, were grown until the cultures reached an optical density of approximately 0.5 at 580 nm. Protein expression was then induced by adding 1 mM isopropyl-β-D-thiogalactopyranoside (IPTG) to the culture medium. Recombinant oxygen biosensor FluBO, YFP and FbFP were purified as His6-tagged proteins from E. coli BL21(DE3) after 16 h of expression at 37°C. Purification was carried out by using HisTag Superflow Hightrap gravityFlow cartridges (Qiagen, Hilden, Germany) and peristaltic pump P-1 (GE Healthcare Europe GmbH, Munich, Germany) under standard operation conditions as described by the manufacturers. The purified proteins were stored at 4°C in protein storage solution containing 10 mM NaCl, 10 mM NaH2PO4, pH 8.0.
Spectral analysis and fluorometry
The absorption and fluorescence properties of FbFP, YFP and FluBO were determined in 10 mM NaCl and 10 mM sodium phosphate buffer pH 8.0 on a UV-2450 absorption spectrophotometer (Shimadzu Europa GmbH, Duisburg, Germany) and a QM-4 fluorescence spectrophotometer (PTI, Ford, West Sussex, UK), the latter with spectral correction for detector and emission monochromator transmission. The absorption spectra were recorded by scanning from 330 nm to 620 nm; the emission spectra were recorded at excitation wavelengths of 380 nm or 440 nm. When emission spectra were measured the maximum absorption of the samples did not exceed 0.15. In the case of excitation/emission (2D) scans, fluorescence excitation was determined from 300 nm to 600 nm in 5 nm steps by using the PerkinElmer LS50B Luminescence Spectrometer (Wellesley, MA, USA) at 22°C. Here, the fluorescence emission spectra were detected from 450 nm to 595 nm wavelength.
Thrombin protease cleavage
For the thrombin protease mediated cleavage of the FluBO fusion protein, 10 μg of purified protein and one unit of thrombin protease (Novagen, distributed by Merck KGaA, Darmstadt, Germany) were diluted in 50 μl thrombin cleavage buffer. After incubation at room temperature for 16 hours an aliquot of 10 μl was used for SDS-PAGE analysis to corroborate the cleavage of the YFP-FbFP fusion by thrombin protease. SDS-PAGE gels were prepared using standard methods (NuPAGE, 4 to 12% Bis-Tris gel 1 mm, Invitrogen, Paisley, UK). After SDS-PAGE separation, proteins were transferred to a PVDF membrane (Bio-Rad Laboratories GmbH, Munich, Germany) by Western blotting (NuPAGE, Invitrogen, Paisley, UK). Subsequently, YFP and FbFP proteins were detected using the respective antibodies (rabbit anti-GFP, BD Biosciences Erembodegem, Belgium; YtvA-specific antiserum). To detect the YFP-FbFP fusion protein FluBO, a His-tag antibody (Anti-His-HRP antibody, Invitrogen, Paisley, UK) was used. Immunodetected proteins were visualized using a CCD camera-based chemiluminescence detection system (Stella imaging system with AIDA advanced image data analyzer, Raytest, Straubenhardt, Germany). Fluorescence emission of FPs was determined by using the PerkinElmer LS50B Luminescence Spectrometer at 22°C as described above.
Fluorescence lifetime imaging and analysis
The E. coli cells expressing FbFP, YFP or FluBO were placed in suspension between two glass slides on the stage of an upright fluorescence microscope (BX50WI; Olympus Optical, Tokyo, Japan) and observed through a 60 × water-immersion objective (NA = 0.9; Olympus Optical). Fluorescence was excited with ca. 100 fs light pulses (λexc = 760 nm) applied at sufficient intensity to generate two-photon excitation. Light pulses were generated at a frequency of 80 MHz by a mode-locked Titan-Sapphire laser (MaiTai DeepSee; output power > 2W; Newport, Spectra Physics Irvine, CA, USA). The laser light was directed through the lens onto the E. coli at reduced power (ca. 5 mW) and scanned over the sample using a beam scanner (TILL Photonics, Munich, Germany). Fluorescence was recorded by a photomultiplier (PMT-100, Becker & Hickl, Berlin, Germany) using appropriate filters for detecting the FbFP fluorescence of FbFP and FluBO (short pass filter 500 nm; λobs < 500 nm; Omega Optical, Brattleboro, VT, USA) and the YFP fluorescence of YFP and FluBO (band pass filter 535 +/- 13 nm; Omega Optical). Fluorescence lifetime imaging was performed using electronics for time-correlated single photon counting (Simple-Tau 152; Becker & Hickl) and acquisition software (SPCM8.95; Becker & Hickl) for time-correlated single-photon counting as described before [50, 51]. Lifetime images were analyzed using SPCImage 2.97 (Becker & Hickl) by fitting mono- or biexponential model functions to the fluorescence decays in every pixel of the image. The program uses an iterative reconvolution of the exponential function with an instrument response function and a least square algorithm for finding parameters of the exponential function for a satisfactory fit.
Characterization of the oxygen biosensor using the BioLector microbioreactor system
All cultivations were carried out in sterile black 48-well microtiter plates (Flowerplate, m2p-labs, Aachen, Germany) in the microcultivation and on-line monitoring system BioLector (m2p-labs, Aachen, Germany) [36, 37]. These microtiter plates allow higher oxygen transfer rates as compared to conventional microtiter plates . Furthermore, each well is equipped with an optode for the determination of DOT (% air saturation) via fluorescence measurement. The microtiter plates were sealed with sterile gas-permeable adhesive seals (Thermo Scientific, Waltham, MA, USA), ensuring sterile conditions in the wells and allowing good ventilation. The following conditions were applied for all cultivations in the BioLector: temperature 30°C, total filling volume per well: 600 μl, shaking diameter: 3 mm, relative humidity in the incubation chamber: 80%. For all cultivations Overnight Express™ Instant TB medium (Novagen, distributed by Merck KGaA, Darmstadt, Germany) was used. Cultures were inoculated from pre-cultures to an initial OD600 of 0.1. Pre-cultures were made in 250 ml flasks under the following conditions: inoculation from a cryoculture to yield an initial OD600 of 0.1, temperature 37°C, total filling volume 10 ml of TB medium, shaking diameter 50 mm, shaking frequency 350 rpm, growth overnight to stationary phase. For the batch cultivation in the BioLector the shaking frequency was 800 rpm. For DOT shift experiments the initial shaking frequency was 600 rpm, which was increased to 700 rpm after 11.2 h and 800 rpm after 12.2 h. The biomass (I-Io) was measured via scattered light intensity at an excitation wavelength of 620 nm. Donor (FbFP) and acceptor (YFP) fluorescence were excited at 380 nm and emission was detected at 492 nm and 532 nm, respectively. DOT was measured at an excitation wavelength of 505 nm and an emission wavelength of 590 nm. The measuring cycle was 10 minutes for the batch cultivation and 4.5 minutes for the DOT shift experiment. Parallel fermentation experiments in the microwells were performed in triplicates. These results were in excellent agreement. Maximum time shifts of 10 minutes occurred due to unavoidable slight differences of cell density at the beginning of the cultivation. Therefore, a representative set of data from these three parallel independent measurements is shown instead of the corresponding mean values.
For the calibration of the optode-derived DOT signal and the FRET signal we considered the derivation of the YFP-FbFP ratio over the time (d (ratio)/d (t)). In this way, it is possible to also follow decreasing DOT values. To reduce the influence of measuring errors for the calculation of d (ratio)/d (t) the curves for the FluBO fluorescence ratio were fitted to the raw data. This was done in MATLAB 7.11 (The MathWorks, Inc., Natick, MA, USA) applying a smoothing spline with a smoothing parameter of 0.999. In the used medium the value of 100% DOT corresponds to an oxygen concentration of 0.24 mmol/l at 30°C and an atmospheric pressure of 100 kPa.
dissolved oxygen tension
flavin-binding fluorescent protein
fluorescence lifetime imaging
fluorescent protein-based biosensor for oxygen
Förster resonance energy transfer
green fluorescent protein
sodium dodecyl sulfate polyacrylamide gel electrophoresis
yellow fluorescent protein.
Aragonés J, Fraisl P, Baes M, Carmeliet P: Oxygen sensors at the crossroad of metabolism. Cell Metab. 2009, 9: 11-22. 10.1016/j.cmet.2008.10.001.
Ward JPT: Oxygen sensors in context. Biochim Biophys Acta. 2008, 1777: 1-14. 10.1016/j.bbabio.2007.10.010.
Ernst JF, Tielker D: Responses to hypoxia in fungal pathogens. Cell Microbiol. 2009, 11: 183-190. 10.1111/j.1462-5822.2008.01259.x.
Hassett DJ, Sutton MD, Schurr MJ, Herr AB, Caldwell CC, Matu JO: Pseudomonas aeruginosa hypoxic or anaerobic biofilm infections within cystic fibrosis airways. Trends Microbiol. 2009, 17: 130-138. 10.1016/j.tim.2008.12.003.
Rustad TR, Sherrid AM, Minch KJ, Sherman DR: Hypoxia: a window into Mycobacterium tuberculosis latency. Cell Microbiol. 2009, 11: 1151-1159. 10.1111/j.1462-5822.2009.01325.x.
Schobert M, Jahn D: Anaerobic physiology of Pseudomonas aeruginosa in the cystic fibrosis lung. Int J Med Microbiol. 2011, 300: 549-556.
Schobert M, Tielen P: Contribution of oxygen-limiting conditions to persistent infection of Pseudomonas aeruginosa. Future Microbiol. 2010, 5: 603-621. 10.2217/fmb.10.16.
Eltzschig HK, Carmeliet P: Hypoxia and inflammation. N Engl J Med. 2011, 364: 656-665. 10.1056/NEJMra0910283.
Brown JM: Exploiting the hypoxic cancer cell: mechanisms and therapeutic strategies. Mol Med Today. 2000, 6: 157-162. 10.1016/S1357-4310(00)01677-4.
Brown JM: Tumor hypoxia in cancer therapy. Methods Enzymol. 2007, 435: 297-321.
Gordan JD, Simon MC: Hypoxia-inducible factors: central regulators of the tumor phenotype. Curr Opin Genet Dev. 2007, 17: 71-77. 10.1016/j.gde.2006.12.006.
Lu X, Kang Y: Hypoxia and hypoxia-inducible factors: master regulators of metastasis. Clin Cancer Res. 2010, 16: 5928-5935. 10.1158/1078-0432.CCR-10-1360.
Coates JD, Anderson RT: Emerging techniques for anaerobic bioremediation of contaminated environments. Trends Biotechnol. 2000, 18: 408-412. 10.1016/S0167-7799(00)01478-5.
Karakashev D, Thomsen AB, Angelidaki I: Anaerobic biotechnological approaches for production of liquid energy carriers from biomass. Biotechnol Lett. 2007, 29: 1005-1012. 10.1007/s10529-007-9360-3.
Löffler FE, Edwards EA: Harnessing microbial activities for environmental cleanup. Curr Opin Biotechnol. 2006, 17: 274-284. 10.1016/j.copbio.2006.05.001.
McKinlay JB, Harwood CS: Photobiological production of hydrogen gas as a biofuel. Curr Opin Biotechnol. 2010, 21: 244-251. 10.1016/j.copbio.2010.02.012.
Fercher A, O'Riordan TC, Zhdanov AV, Dmitriev RI, Papkovsky DB: Imaging of cellular oxygen and analysis of metabolic responses of mammalian cells. Methods Mol Biol. 2010, 591: 257-273. 10.1007/978-1-60761-404-3_16.
Lee YE, Smith R, Kopelman R: Nanoparticle PEBBLE sensors in live cells and in vivo. Annu Rev Anal Chem (Palo Alto Calif). 2009, 2: 57-76. 10.1146/annurev.anchem.1.031207.112823.
O'Riordan TC, Zhdanov AV, Ponomarev GV, Papkovsky DB: Analysis of intracellular oxygen and metabolic responses of mammalian cells by time-resolved fluorometry. Anal Chem. 2007, 79: 9414-9419. 10.1021/ac701770b.
Delvigne F, Boxus M, Ingels S, Thonart P: Bioreactor mixing efficiency modulates the activity of a prpoS::GFP reporter gene in E. coli. Microb Cell Fact. 2009, 8: 15-10.1186/1475-2859-8-15.
Garcia JR, Cha HJ, Rao G, Marten MR, Bentley WE: Microbial nar-GFP cell sensors reveal oxygen limitations in highly agitated and aerated laboratory-scale fermentors. Microb Cell Fact. 2009, 8: 6-10.1186/1475-2859-8-6.
Takahashi E, Sato M: Impact of intracellular diffusion of oxygen in hypoxic sensing. Adv Exp Med Biol. 2011, 701: 301-306. 10.1007/978-1-4419-7756-4_40.
Takahashi E, Takano T, Nomura Y, Okano S, Nakajima O, Sato M: In vivo oxygen imaging using green fluorescent protein. Am J Physiol Cell Physiol. 2006, 291: C781-787. 10.1152/ajpcell.00067.2006.
Takahashi E, Takano T, Numata A, Hayashi N, Okano S, Nakajima O, Nomura Y, Sato M: Genetic oxygen sensor: GFP as an indicator of intracellular oxygenation. Adv Exp Med Biol. 2005, 566: 39-44. 10.1007/0-387-26206-7_6.
Li IT, Pham E, Truong K: Protein biosensors based on the principle of fluorescence resonance energy transfer for monitoring cellular dynamics. Biotechnol Lett. 2006, 28: 1971-1982. 10.1007/s10529-006-9193-5.
Newman RH, Fosbrink MD, Zhang J: Genetically encodable fluorescent biosensors for tracking signaling dynamics in living cells. Chem Rev. 2011, 111: 3614-3666. 10.1021/cr100002u.
Wang H, Nakata E, Hamachi I: Recent progress in strategies for the creation of protein-based fluorescent biosensors. Chembiochem. 2009, 10: 2560-2577. 10.1002/cbic.200900249.
Shaner NC, Steinbach PA, Tsien RY: A guide to choosing fluorescent proteins. Nat Methods. 2005, 2: 905-909. 10.1038/nmeth819.
Tsien RY: The green fluorescent protein. Annu Rev Biochem. 1998, 67: 509-544. 10.1146/annurev.biochem.67.1.509.
Drepper T, Huber R, Heck A, Circolone F, Hillmer AK, Buchs J, Jaeger KE: Flavin mononucleotide-based fluorescent reporter proteins outperform green fluorescent protein-like proteins as quantitative in vivo real-time reporters. Appl Environ Microbiol. 2010, 76: 5990-5994. 10.1128/AEM.00701-10.
Drepper T, Eggert T, Circolone F, Heck A, Krauss U, Guterl JK, Wendorff M, Losi A, Gärtner W, Jaeger KE: Reporter proteins for in vivo fluorescence without oxygen. Nat Biotechnol. 2007, 25: 443-445. 10.1038/nbt1293.
Frommer WB, Davidson MW, Campbell RE: Genetically encoded biosensors based on engineered fluorescent proteins. Chem Soc Rev. 2009, 38: 2833-2841. 10.1039/b907749a.
Ibraheem A, Campbell RE: Designs and applications of fluorescent protein-based biosensors. Curr Opin Chem Biol. 2010, 14: 30-36. 10.1016/j.cbpa.2009.09.033.
Palmer AE, Qin Y, Park JG, McCombs JE: Design and application of genetically encoded biosensors. Trends Biotechnol. 2011, 29: 144-152. 10.1016/j.tibtech.2010.12.004.
Kremers G-J, Goedhart J, van Munster EB, Gadella TWJ: Cyan and yellow super fluorescent proteins with improved brightness, protein folding, and FRET Förster radius. Biochemistry. 2006, 45: 6570-6580. 10.1021/bi0516273.
Kensy F, Zang E, Faulhammer C, Tan RK, Büchs J: Validation of a high-throughput fermentation system based on online monitoring of biomass and fluorescence in continuously shaken microtiter plates. Microb Cell Fact. 2009, 8: 31-10.1186/1475-2859-8-31.
Samorski M, Müller-Newen G, Büchs J: Quasi-continuous combined scattered light and fluorescence measurements: a novel measurement technique for shaken microtiter plates. Biotechnol Bioeng. 2005, 92: 61-68. 10.1002/bit.20573.
Arosio D, Ricci F, Marchetti L, Gualdani R, Albertazzi L, Beltram F: Simultaneous intracellular chloride and pH measurements using a GFP-based sensor. Nat Methods. 2010, 7: 516-518. 10.1038/nmeth.1471.
Esposito A, Gralle M, Dani MA, Lange D, Wouters FS: pHlameleons: a family of FRET-based protein sensors for quantitative pH imaging. Biochemistry. 2008, 47: 13115-13126. 10.1021/bi8009482.
Tantama M, Hung YP, Yellen G: Imaging intracellular pH in live cells with a genetically encoded red fluorescent protein sensor. J Am Chem Soc. 2011, 133: 10034-10037. 10.1021/ja202902d.
Yano T, Oku M, Akeyama N, Itoyama A, Yurimoto H, Kuge S, Fujiki Y, Sakai Y: A novel fluorescent sensor protein for visualization of redox states in the cytoplasm and in peroxisomes. Mol Cell Biol. 2010, 30: 3758-3766. 10.1128/MCB.00121-10.
Minton AP: How can biochemical reactions within cells differ from those in test tubes?. J Cell Sci. 2006, 119: 2863-2869. 10.1242/jcs.03063.
van Eunen K, Bouwman J, Daran-Lapujade P, Postmus J, Canelas AB, Mensonides FI, Orij R, Tuzun I, van den Brink J, Smits GJ, van Gulik WM, Brul S, Heijnen JJ, de Winde JH, de Mattos MJ, Kettner C, Nielsen J, Westerhoff HV, Bakker BM: Measuring enzyme activities under standardized in vivo-like conditions for systems biology. Febs J. 2010, 277: 749-760. 10.1111/j.1742-4658.2009.07524.x.
Dmitriev RI, Ropiak HM, Yashunsky DV, Ponomarev GV, Zhdanov AV, Papkovsky DB: Bactenecin 7 peptide fragment as a tool for intracellular delivery of a phosphorescent oxygen sensor. FEBS J. 2010, 277: 4651-4661. 10.1111/j.1742-4658.2010.07872.x.
Fercher A, Ponomarev GV, Yashunski D, Papkovsky D: Evaluation of the derivates of phosphorescent Pt-coproporphyrin as intracellular oxygen-sensitive probes. Anal Bioanal Chem. 2010, 396: 1793-1803. 10.1007/s00216-009-3399-z.
Elowitz MB, Surette MG, Wolf PE, Stock J, Leibler S: Photoactivation turns green fluorescent protein red. Curr Biol. 1997, 7: 809-812. 10.1016/S0960-9822(06)00342-3.
Hanahan D: Studies on transformation of Escherichia coli with plasmids. J Mol Biol. 1983, 166: 557-580. 10.1016/S0022-2836(83)80284-8.
Ormö M, Cubitt AB, Kallio K, Gross LA, Tsien RY, Remington SJ: Crystal structure of the Aequorea victoria green fluorescent protein. Science. 1996, 273: 1392-1395. 10.1126/science.273.5280.1392.
Katzke N, Arvani S, Bergmann R, Circolone F, Markert A, Svensson V, Jaeger K-E, Heck A, Drepper T: A novel T7 RNA polymerase dependent expression system for high-level protein production in the phototrophic bacterium Rhodobacter capsulatus. Protein Exp Purif. 2010, 69: 137-146. 10.1016/j.pep.2009.08.008.
Gilbert D, Franjic-Würtz C, Funk K, Gensch T, Frings S, Möhrlen F: Differential maturation of chloride homeostasis in primary afferent neurons of the somatosensory system. Int J Dev Neurosci. 2007, 25: 479-489. 10.1016/j.ijdevneu.2007.08.001.
Kaneko H, Putzier I, Frings S, Kaupp UB, Gensch T: Chloride accumulation in mammalian olfactory sensory neurons. J Neurosci. 2004, 24: 7931-7938. 10.1523/JNEUROSCI.2115-04.2004.
Funke M, Diederichs S, Kensy F, Müller C, Büchs J: The baffled microtiter plate: increased oxygen transfer and improved online monitoring in small scale fermentations. Biotechnol Bioeng. 2009, 103: 1118-1128. 10.1002/bit.22341.
The authors would like to thank Prof. Dr. Martina Pohl and Dr. Achim Heck for helpful discussion.
The authors declare competing financial interests. Parts of the results described in this publication have been included into a patent application (Circolone F., Drepper T., Endres S. Heck A., Jaeger K.-E., Potzkei J. (2010); Patent number DE 10 2010 037 001). The oxygen biosensor FluBO will be commercialized by evocatal GmbH, Germany.
TD developed the idea for the project and designed the experiments concerning the construction and characterization of the biosensor. JP designed and performed the experiments concerning the construction and characterization of the biosensor. MK and JB designed and performed the BioLector experiments. TG designed and performed the FLIM experiments. JP, MK, TG, TD and JB developed the method for data processing and analysis. TD and K-EJ wrote the paper. All authors read and approve the final manuscript.
Janko Potzkei, Martin Kunze, Thomas Drepper contributed equally to this work.
Electronic supplementary material
Additional file 1: . Yellow, YFP sequence; black, linker sequence; green, FbFP sequence; grey, His6 tag sequence. (DOC 39 KB)
Additional file 2: Fluorescence spectra of FluBO before and after proteolysis with thrombin. Fluorescence spectra of FluBO before and after proteolysis with thrombin. The fluorescence emission spectra of purified FPs (5 μM) were recorded at an excitation wavelength of 380 nm. Complete thrombin-catalyzed proteolysis of FluBO was demonstrated by SDS-PAGE (inset), purified FPs were used as standards. (JPEG 448 KB)
Additional file 3: . The figure shows the normalized YFP fluorescence intensity at increasing pH values. Purified YFP was adjusted to an absorption of 0.05. The effect of the pH value on YFP fluorescence was measured in citric acid/disodium phosphate buffer with pH values ranging form 4.0 to 7.0. The pKa value is marked by doted lines. (JPEG 156 KB)
Additional file 4: Data from parallel fermentation experiments. The figure shows the individual results from three independent batch cultivation experiments using the BioLector. The development of biomass (black), dissolved oxygen tension (blue) and cyan-to-yellow fluorescent ratio (emYFP/emFbFP) of the FRET-based oxygen biosensor (red) was online-monitored in each well of the flowerplate. (JPEG 1 MB)
Authors’ original submitted files for images
Below are the links to the authors’ original submitted files for images.
About this article
Cite this article
Potzkei, J., Kunze, M., Drepper, T. et al. Real-time determination of intracellular oxygen in bacteria using a genetically encoded FRET-based biosensor. BMC Biol 10, 28 (2012). https://doi.org/10.1186/1741-7007-10-28
- Green Fluorescent Protein
- Yellow Fluorescent Protein
- Intracellular Oxygen
- Yellow Fluorescent Protein Fluorescence
- Acceptor Domain