- Question and Answer
- Open Access
Q&A: Array tomography
© Smith et al. 2018
- Published: 6 September 2018
Array tomography encompasses light and electron microscopy modalities that offer unparalleled opportunities to explore three-dimensional cellular architectures in extremely fine structural and molecular detail. Fluorescence array tomography achieves much higher resolution and molecular multiplexing than most other fluorescence microscopy methods, while electron array tomography can capture three-dimensional ultrastructure much more easily and rapidly than traditional serial-section electron microscopy methods. A correlative fluorescence/electron microscopy mode of array tomography furthermore offers a unique capacity to merge the molecular discrimination strengths of multichannel fluorescence microscopy with the ultrastructural imaging strengths of electron microscopy. This essay samples the first decade of array tomography, highlighting applications in neuroscience.
Array tomography (AT) is a versatile microscopy method that offers superlative opportunities to explore cell and tissue architectures in three dimensions. It is well suited to seamless imaging of large tissue volumes in extremely fine structural and high molecular detail, positioning the method nicely for emerging post-transcriptomic tissue biology applications. A fluorescence microscopy AT mode (FM-AT) delivers volumetric resolution and molecular marker multiplexing highly superior to traditional fluorescence microscopies, while an electron microscopy AT mode (EM-AT) readily captures three-dimensional ultrastructure at size scales that would require prohibitive effort using traditional serial-section EM methods. Where AT is entirely unique, however, is in supporting a “voxel-conjugate” combination of the fluorescence and electron microscopy modes (FM/EM-AT), where three-dimensional light and electron images are acquired in essentially perfect volumetric register. These attributes establish AT as an ideal choice for the most demanding analyses of diverse cellular architectures within mature and developing tissues, including brain. This essay will draw examples mainly from neuroscience, but AT methods are also finding many cell and tissue biology applications outside of neuroscience [1–13].
Additional file 1: Movie S1. Fluorescence (FM-AT) and electron (EM-AT) images of a single mouse cortex array section, overlaid in pixel-precise register illustrating combination of multichannel FM of a large area (approximately 0.4 × 1.0 mm, spanning all six cortical layers) with EM sampling of a smaller subfield (unpublished Allen Institute data). (Procedures as schematized in arrow track 3, Fig. 2. The rationale for registering a large FM-AT field with a much smaller EM-AT field is discussed in the text section “What limits AT specimen size”). In this video rendering, the field of view gradually zooms 200-fold into a very small subfield in layer 5. At the higher zooms, it is evident that synaptic protein (PSD95, GluN1, VGluT1, Synapsin, GAD2 and Gephyrin), nuclear DNA (DAPI), myelin (MBP), and GABA markers align with EM images as expected from current biological models of mammalian cortex and synapses. Colors representing ten channels of molecular fluorescence are modulated periodically in this video to better accommodate the limitation of human color vision to (at most) three discrete color channels. The specimen samples VISp cortex of a transgenic mouse in which expression of a fluorescent protein (TdTomato) was driven mainly in layer 4 pyramidal cells  (MOV 117816 kb)
Additional file 2: Movie S2. Visualization of an eight-channel FM-AT volume of mouse somatosensory cortex illustrating: a results of high-order sequential marker multiplexing (as schematized in Fig. 2 and discussed text sections “What makes AT especially suitable for sequential multiplexing”); and b the high volumetric resolution of FM-AT (as discussed in “What limits FM-AT resolution” section). This volume (h, w, d = 130 × 90 × 3.5 um) was sampled from a Thy1-H-YFP transgenic mouse (unpublished data courtesy of Kristina Micheva). Specimen and all methods are as previously described . Eight colors representing synaptic (Synapsin, PSD95, GAD), neuron type (Thy1-H, parvalbumin, calbindin), and organelle (DAPI, tubulin) markers are modulated in separate groups to accommodate limitation of human color vision to three discrete channels (MOV 104130 kb)
Additional file 3: Movie S3. “Machinery of Mind” renders an FM-AT volume image of mouse somatosensory cortex to evoke the beautiful intricacies typical of the synaptic networks that endow all animal nervous systems with their astonishing functional capabilities. This video is best appreciated when played with sound (original musical score and performance by Catherine Rose Smith) over a full-range audio system or good headphones, turned up loud. This is a revised version of supplemental video #1 from Micheva, et al.  (Copyright (2010), with permission from Elsevier), enhanced by the addition of a new prolog sequence, based on an MRI atlas dataset  to indicate the relationship of the FM-AT volume sampled (approximately 1.5 mm by 0.5 mm wide by 12 μm deep) to the whole mouse brain. A subset of glutamatergic neurons (mostly layer 5 pyramidal cells) are rendered in green, reflecting expression of YFP in the Thy-1H-YFP mouse from which the specimen was obtained. The red puncta evident in this FM-AT visualization correspond to individual synapses: approximately eight million can be resolved within the rendered volume. The blue filaments represent microtubule bundles and are visible mainly in the dendrites of non-YFP expressing neurons. Details of the FM-AT volume capture are as described in . Briefly, the FM-AT volume data were captured by imaging a single ribbon of 60 serial sections, each cut 200 nm thick from a block of LR-White embedded tissue and labeled by post-embedding immunostaining. The FM-AT volume extends from the pial surface through all six layers of cortex and subcortical white matter into the striatum. This dataset comprises 3.7 billion voxels and three color planes (each one byte wide: YFP, green; anti-Synapsin I, red; anti-alpha-tubulin, blue). (MOV 579726 kb)
The term “array tomography” was introduced by a 2007 neuroscience publication , but earlier and contemporaneous writings presage individual fluorescence and electron microscopy elements of AT [18–22]. The terminology is straightforward: “array” refers to arrangement of serial sections in spatial array on a planar solid surface, while “tomography” alludes to the capture of three-dimensional structure from two-dimensional image “slices” (fr. Greek “tomos”, slice). Confusion may result, however, from the widespread use of “computed axial tomography” (CAT) in reference to a form of three-dimensional X-ray imaging widely used in clinical radiology, and of “electron tomography” (ET) in reference to an ultra-high-resolution form of three-dimensional electron microscopy. For both CAT and ET, volumetric image “slices” are computed from projection images acquired from multiple angles and no physical slicing is usually involved [23, 24]. Unlike CAT and ET, AT does not generally involve transforming rotational projections.
As defined above, AT might conceivably apply to all forms of serial-section microscopy, including serial-section transmission electron microscopy (ssTEM) [25, 26]. The AT terminology is restricted here, however, to arrays placed on stable solid substrates such as glass coverslips, flexible tape, or silicon wafers, as opposed to the open grid slots or delicate, ultrathin electron-transparent support films required for ssTEM imaging. Physical stability of the array substrate is essential to several distinctive AT benefits, such as specimen stability during the repeated solution changes necessary for sequential multiplexing. As electron absorption by a substantial solid substrate precludes easy use of transmission electron microscopy (TEM), scanning electron microscopy (SEM) is most often used for EM-AT because backscattered or secondary electron images can be acquired in a reflection mode. Though TEM offers the ultimate in electron microscopic resolution, SEM resolution is more than adequate for a wide range of cell and tissue science applications. That said, one recently introduced hybrid variant of AT, called AT-TEM, uses a film transfer method to conjoin the stability advantages of fluorescence AT imaging on a solid substrate with the resolution and speed advantages of subsequent TEM imaging [27, 28].
The use of FM-AT (Figs. 2, 3 and 4) should be considered for volumetric fluorescence imaging of fixed tissue specimens whenever there is need for very high resolution, high-order molecular multiplexing and/or rigorously depth-independent quantification of fluorescence signal intensities. Use of EM-AT (Figs. 2 and 5) offers perhaps the most convenient approach to volumetric electron microscopy available. Use of FM/EM-AT (Figs. 2, 6 and 7) offers unique opportunities to combine the molecular discrimination strengths of fluorescence microscopy with the unrivaled structural resolution of electron microscopy [1, 2, 6, 11, 14, 15, 17, 27–37]. While other approaches to correlative light and electron microscopy of individual specimens have proven extremely useful , AT alone offers a path to perfect registration of fluorescence and SEM voxels over extended volumes. The unique ability of conjugate FM/EM-AT to unify molecular and ultrastructural views of neural network architectures sets a standard for the emerging fields of synaptomics and connectomics. The analysis of diverse cortical synapse populations (as illustrated in Figs. 3, 4, 5, 6 and 7) provides examples of applications benefitting from each of the major AT strengths.
AT confers an additional special advantage for quantitative fluorescence imaging of brain specimens from older animals (including all adult humans), which are often suffused with brightly autofluorescent lipofuscin deposits. With standard fluorescence microscopy methods, out-of-focus flare from lipofuscin usually obscures nearby tissue features and confounds fluorescence interpretation and quantification . The extremely high Z-axial resolution resulting from AT ultrathin physical sectioning eliminates such lipofuscin interference decisively (Fig. 4e) and may enable improved fluorescence analysis of the intimate cellular milieu in which lipofuscin forms in aging brains.
By definition, all forms of AT require serial sections to be cut and transferred to a solid substrate for imaging. Cutting the very thin sections necessary for high-resolution AT requires in turn that tissue first be fixed, dehydrated, and embedded in a hard, cross-linked polymer resin matrix. Years of transmission electron microscopy (TEM) and immuno-TEM practice have resulted in the development and optimization of a variety of alternative fixation, dehydration, and embedding materials and methods , but it is necessary to consider various tradeoffs in choosing amongst these alternatives.
Mild chemical fixation, e.g., by formaldehyde alone, is generally preferable for preservation of immunoreactivity but compromises preservation of fine ultrastructural details. More stringent chemical fixatives, e.g., glutaraldehyde and/or osmium, better preserve ultrastructure but compromise immunoreactivity. Dehydration methods also strongly influence tradeoffs between preservation and immunoreactivity, with freeze substitution methods (where the specimen is frozen and water is replaced by organic solvent at very low temperatures) generally yielding superior results but requiring more complex procedures and equipment in comparison with room-temperature solvent replacement methods. The choice of embedding resin chemistry also entails a tradeoff, with acrylic embedding resins (such as LR White and Lowicryls) offering much better antibody access for immunofluorescence while epoxy resins generally yield superior EM image quality. While ultrathin sections can be cut by resin-free cryosectioning methods, which might avoid structure–immunoreactivity tradeoffs, no applications to AT have yet appeared, reflecting the imposing difficulty of serial cryosection production and collection. Each of the other fixation, dehydration, and resin-embedding methods mentioned here has been employed for AT, with choices amongst these diverse methods being driven primarily by the project goals and tradeoffs outlined above [5, 17, 31, 41–43]. In choosing a tissue preparation method for any specific AT application, the established literatures from electron and correlative light/electron microscopy practice will provide much additional useful guidance [38, 40, 44–48].
The diamond-knife ultramicrotome, refined over many decades of use for transmission electron microscopy (TEM; Fig. 1b), accomplishes the basic, automated cutting of ultrathin serial sections easily and reliably. All forms of AT described so far employ this standard instrument, which automatically cuts either individual, free-floating sections or continuous “ribbons” of serial sections (Fig. 1c–e) onto the surface of water held in a small pool just behind the diamond knife edge. These very thin and delicate sections or ribbons must then be transferred from the water surface to a solid substrate. Simple manual means, usually involving an “eyelash” tool to move section ribbons on the water surface, have sufficed for such transfer in small-scale projects, but applications requiring the imaging of larger tissue volumes, such as those required for the analysis of extended synaptic networks, have motivated development of higher-throughput, automated means of water-to-solid section transfer as necessary to build large-scale arrays.
A variety of materials have been used as AT array substrates. Early AT substrates were traditional “subbed” histology slides—standard glass microscope slides coated with a layer of hardened gelatin to promote section adhesion . While serviceable for less demanding AT applications, this substrate is limited by less-than-ideal optical properties, mediocre section adhesion, and instability under the electron beam. Improved optics can be obtained by adhering the sections or ribbons directly to a precision coverslip, rather than onto a slide, but the gelatin layer remains problematic. Much improved FM-AT and EM-AT image quality and array stability are now obtained by replacing the gelatin with a thin, transparent layer of evaporated carbon laid down upon a silanized coverslip surface . Flexible polymer substrates, such as carbon-coated Kapton tape, have enabled the use of a simple tape-transport method for automated, high-throughput array section collection [2, 10, 29, 49, 55]. Continuous tapes of array sections may be readily cut into shorter segments and glued to small silicon wafers in multi-row arrays for SEM imaging. Other electron and fluorescence AT methods involve direct adhesion of array sections to silicon wafer surfaces [6, 30].
FM-AT signals may result from immunofluorescence labeling, dye injection, or transgenically expressed fluorescent proteins [6, 17]. Signals from DNA FISH have also been demonstrated . Immunolabeling may be accomplished either prior to resin embedment (pre-embedding) or after embedding and sectioning (post-embedding). Pre-embedding immunolabeling generally offers higher immunolabeling efficiency near the specimen surface, but at the cost of strongly depth-dependent labeling efficiency. Post-embedding labeling (performed after ultrathin sectioning) offers depth-insensitive labeling, while sacrificing molar labeling efficiency and thereby signal-to-noise ratio. Post-embedding immunostaining may also be preferred because of superior preservation of ultrastructure  and superior sequential multiplexing possibilities. The minimal thickness of AT sections greatly facilitates quick and reliable specimen staining by eliminating the need for staining molecules to percolate via binding-restricted diffusion into a thick tissue section. It should be noted, however, that some antibodies that work well for pre-embedding staining do not work as well in post-embedding applications (including AT), presumably because specimen dehydration and resin embedding alter or hide target protein epitopes [40, 58]. It is, therefore, generally advisable to search any available resources [17, 31, 37, 59–61] regarding antibodies that have established efficacy in post-embedding or AT applications, or otherwise be prepared to test multiple antibodies for suitability to such specimens .
For EM-AT, contrast is usually generated by staining cellular membranes and proteins non-specifically with heavy metals such as osmium, lead, or uranium. Again, staining may be accomplished either pre-embedding or post-embedding, but pre-embedding metal staining can be considered only when no subsequent immunolabeling is intended, as the methods are incompatible. Pre-embedding metal staining can provide excellent results with small tissue samples, but results are often inconsistent with larger specimens. Post-embedding staining for EM-AT avoids any depth-dependent stain variations and can be carried out following the conclusion of FM-AT imaging, providing the essential basis for conjugate FM/EM-AT image acquisition.
FM-AT images may be acquired by standard widefield fluorescence microscopy, by confocal fluorescence microscopy, or by lateral super-resolution modes such as PALM, STORM, STED, or structured illumination. The choice of fluorescence microscopy mode is influenced by tradeoffs between achievable lateral resolution, imaging speed, and the number of fluorescence “color” channels accessible in a single stain-image-wash round. Straight widefield fluorescence offers a very attractive combination of low instrument cost, very high acquisition speed, high channel capacity, and high resolution, truly diffraction-limited even when using aberration-prone high NA objectives (as explained in the following section). When sufficiently high fluorescent label density is achievable (e.g., [3, 22, 44, 63]), the lateral super-resolution AT modes offer still higher lateral resolution, though this advantage comes at present with substantial costs in process complexity and speed.
Scanning electron microscopy (SEM) captures EM images for EM-AT and FM/EM-AT. While simpler tungsten-filament SEMs might be used for this purpose, the improvements in acquisition speed and effective resolution with the more complex (and, unfortunately, expensive) field-emission SEM (FE-SEM) are dramatic. Though FE-SEM image acquisition is fast in comparison to tungsten SEMs, it is still very slow, however, in comparison to fluorescence image acquisition on a volume basis. A recently introduced multi-beam SEM (mSEM) promises enormous increases in SEM acquisition speed, though this advantage entails further very large increases in instrument complexity and cost.
For both FM-AT and EM-AT modes, computational automation of microscope mechanical axes and image acquisition is extremely helpful and becomes a virtual necessity when it is desired to image large specimen volumes, where many thousands of multispectral image tiles must be acquired. Focus (and in the case of SEM, stigmation of the electron beam) must be automated and XY stage motors must advance the field of view automatically to image many sections in sequence. When imaging larger specimen volumes, it is generally necessary to extend the microscope’s limited two-dimensional field of view by lateral mosaic stitching of multiple image fields. The automation of multichannel fluorescence acquisition is often accomplished by motorizing filter changers, although some newer solid-state light sources and multi-band filter sets allow faster channel selection by switching excitation bands without moving parts. A fluorescence microscope optimized at the Allen Institute for high-throughput, highly automated staining and imaging of AT arrays is depicted in Fig. 8c. Commercial availability of hardware and software specialized for all AT image acquisition modes now promises to open these powerful but complex AT methodologies to much wider adoption [50, 53, 64, 65].
Fluorescence AT readily yields volumetric resolution much higher than whole-mount, diffraction-limited fluorescence methods, such as wide-field or confocal microscopy. The AT improvement in resolution begins with the fact that AT physical sections are usually much thinner (40–100 nm) than the diffraction limit along the focal (Z) axis (theoretically > 500 nm, even at the highest NAs, but usually further worsened in tissue whole-mounts by optical aberrations). The volumetric resolution of FM-AT thus immediately improves by a factor of 5–10 over the Z-axis diffraction limit. This is significant since Z-axis resolution is always much worse than lateral resolution, and is therefore the Achilles’ heel of normal fluorescence resolution.
Lateral (X-Y) FM-AT resolution is also improved substantially in comparison to whole-mount fluorescence, due to minimization of optical aberrations that otherwise compromise lateral resolution in thick specimens. Since AT sections are very thin and placed exactly at the surface of a precision optical coverslip, the stringent conditions required for truly diffraction-limited resolution by a high-NA oil objective are met exactly . With thicker specimens, resolution-robbing aberrations due to optical inhomogeneities in specimen or mounting medium are very difficult to avoid. Moreover, since array sections are much thinner than the optical depth of focus, noise contributions from out-of-focus specimen elements are completely eliminated. These factors dramatically improve image quality and permit application of the most precise two-dimensional deconvolution methods for optimal two-dimensional image restoration . These reductions in optical aberrations and image noise may boost effective lateral resolution for AT by at least a factor of two in comparison to typical whole-mount fluorescence imaging. These same improvements in basic imaging optics confer substantial benefits even when using super-resolution array imaging methods such as PALM, STORM, or STED, because these methods still benefit in speed and limiting resolution from truly diffraction-limited optics .
AT specimens prepared as a single ribbon array (e.g., Fig. 1) might comprise 50 serial sections, each perhaps 2 mm wide by 0.5 mm long by 100 nm thick, yielding a total specimen volume 2 × 0.5 × 0.005 mm in X, Y, and Z. The logistics of serial sectioning and fluorescence imaging naturally encourage acquisition of such “flattened” specimen cuboids, where the least dimension lies along the Z-axis. The single ribbon volume in this example would comprise 5 nL (nanoliters). A total specimen volume of 5 nL may sound small, but note that one nanoliter of mouse cortical gray matter comprises approximately one million synapses! Many of the synaptomic FM-AT volumes illustrated here and in other publications involve only a fraction of a nanoliter, yet represent very large numbers of reliable single-synapse observations. EM-AT volumes may be even smaller: the spectacular EM-AT images rendered in Fig. 5c resulted from mid-resolution imaging of approximately 25 nL and high-resolution imaging of only 0.08 nL of mouse cortex. The acquisition and processing of AT images at the 5 nL scale with today’s fastest technologies may take a few days (for FM-AT) to several weeks (for EM-AT) and entail the reasonable data storage costs associated with a few to a few dozen terabytes.
For the analysis of larger structures, such as complete neuronal dendrites and local circuits, it may be desired to image much larger volumes, on the order of 1000 nL (i.e., one microliter or 1 cubic millimeter). Larger and thicker high-resolution AT volumes necessarily require collecting and imaging larger numbers (i.e., thousands or tens of thousands) of serial sections collected using some form of automation (a tape collector or a multi-ribbon collection robot like that illustrated in Fig. 8a, b). The upper limits to practical AT specimen size are then essentially economic: the costs in hardware, reagents, image acquisition time, data storage requirements, and image processing and analysis loads. All the recurring costs of AT scale approximately linearly with specimen volume and at present must be considered expensive. Figure 3a, b represent the largest AT volumes so far reported in the literature, comprising 175 nL and 50 nL, respectively. These efforts have required extensive custom microscope engineering and software development and person-years of experimental effort. The acquisition and processing of AT images at the 1000 nL scale with these evolving technologies may take several weeks (for FM-AT) to many months (for EM-AT) and entail the daunting data storage costs presently associated with petabyte scale requirements. The upper limits to practical AT specimen size are thus essentially economic: the costs in hardware, image acquisition time, data storage requirements, and image processing and analysis loads. Thus, imaging at the 1000 nL scale looms as a formidable and expensive challenge today. Nonetheless, assuming continued advances in both imaging tool speed and data processing and storage economics, it is reasonable to imagine that 1000 nL and still larger volumes may become much more practical in coming years.
Large differentials in cost between FM-AT image acquisition (fast and inexpensive) and EM-AT imaging (much slower and more expensive) suggest consideration of hybrid FM/EM sampling strategies for many types of project. The FM/EM-AT modality allows for the imaging of relatively large volumes by FM-AT to be followed by the sparse sampling of smaller subvolumes by EM-AT, capturing many key advantages of both AT modes at reasonable cost. Information gained from the sparse FM/EM-AT volumes can be used to more deeply and rigorously interpret the larger, purely FM-AT volumes. The power of this approach has been demonstrated by recent analyses of mouse CNS synapse populations [27, 28, 31] and it seems likely that buttressing the strengths of FM-AT with sparse EM-AT in this way will help to manage the costs of data acquisition handling associated with many future large-scale AT projects.
Rapid increases in readily accessible computational power, digital storage capacities, and fluorescence molecular assay strategies have kindled substantial interest in sequential multiplexing methods for molecular microscopy. The 2007 introduction of AT  demonstrated sequential multiplexing to read out 11 molecular markers and quantitative stability across six sequential rounds of differential staining and imaging. A 2010 publication  demonstrated quantitative imaging of 18 markers by six sequential rounds of fluorescence imaging.
Sequential molecular multiplexing methods based on in situ sequencing and decoding of DNA-bar-coded in situ hybridization probes or bar-coded antibodies are expected to soon advance multiplexing far beyond present practices for both AT and non-AT molecular imaging [70, 71]. With growing excitement about such “bar coding” methods, however, attention must be paid to fundamental limits. Sequential multiplexing requires repeated rounds of imaging interspersed over time by labeling and rinsing washes, with some form of probe elution or bleaching, followed by imaging processing steps to bring images from the sequential rounds into spatial register. Any physical instability during sequential imaging rounds will complicate cross-round image registration and place limits on multiplexing possibilities. Three-dimensional specimens, even when reinforced by polymer gel fixation, are subject to deformation over time and especially when subjected to different staining, wash, or elution solutions. Such instabilities tend to worsen when tissue proteins and lipids are replaced by water, as in the various tissue clearing and expansion methods now coming into use. Because deformations of thick tissue samples are likely to include anisotropic and unpredictable components, imperfect registration of sequential images will limit multiplex image interpretation and compromise prospects for single-molecule multiplexing (e.g., for in situ sequencing or probe decoding), particularly at high target labeling densities. The extreme physical stability of AT specimens, where resin-embedded ultrathin sections are tightly bonded to a solid substrate, decisively eliminates such registration difficulties.
Fluorescence AT offers unique opportunities to quantify fluorescence signals independent of depth within a tissue specimen. When imaging whole-mount tissue specimens, fluorescence readout tends to fall off with increasing focal depth due to increasing light scattering, absorption, and optical aberrations. When whole-mount specimens are labeled by immunostaining, quantitative analysis is further compromised by stain reagent diffusion limits. All such effects are eliminated by AT physical sectioning, where each section—regardless of depth with the original specimen—is stained and imaged identically in a planar format.
The processing of AT images usually begins with flat-field correction  to correct for variations in image brightness across individual image tiles due to microscope illumination or detection characteristics. Images of ultrathin AT sections are then ideal for restoration of high spatial frequencies (i.e., fine image details) by optimal deconvolution methods such as Richard-Lucy that assume a planar specimen geometry [32, 67]. Following such two-dimensional image restoration steps, precise computational alignment of two-dimensional serial section images to reconstruct a three-dimensional volume image is fundamental to all forms of AT [73–75]. For larger volumes, it is also usually necessary to stitch together multiple camera image tiles to compose seamless two-dimensional image section mosaics. Small geometric distortions typically occur during sectioning and section collection and generally require non-rigid transformation of individual tiles and sections to achieve sub-pixel-accurate alignment. When multiple images acquired across multiple image acquisition sessions (as in Fig. 1b) or across multiple microscopes (as in Fig. 1d) rigid and/or non-rigid transformations may also be required to register all two-dimensional image planes into a common pixel space.
As with all forms of serial-section microscopy, it is occasionally necessary to deal gracefully with section defects such as wrinkles, folds, tears, surface contamination, or even the occasional missing section. Since such defects are ordinarily rare and sparse, the data loss per se is usually not overly troublesome, but their presence may perturb high-quality alignment unless recognized and corrected. Finally, when integrating fluorescence and electron AT modalities for FM/EM-AT, image processing must deal efficiently with the wide differential of pixel sizes appropriate to fluorescence (~ 100 × 100 nm) and electron (~ 3 × 3 nm) image acquisition. Moreover, even though AT makes registration of FM and EM images in the Z dimension trivial, useful registration of FM and EM images in X and Y axes must be accurate to a scale set by the very small EM pixel size and may require highly specialized image alignment methods [31, 76]. All of these requirements add up to substantial demands for computational resources, such that cluster or cloud computing and web-based solutions become most appropriate [77–79]. Computational automation of complex AT image processing workflows becomes a virtual necessity as AT imaging is scaled to larger tissue volumes.
Specific protocols for analysis of AT images naturally depend on the nature of the specimen and the biological question addressed. The prevalent AT application to date has been in synaptomics, where AT is prized for reliable resolution, detection, and measurement of synapses crowded into brain tissue context [17, 27, 28, 30, 31, 35, 55, 63, 80–98]. Figures 3, 4, 5, 6 and 8 illustrate a small selection of results of synaptomic and other published analysis protocols; the publications cited in the corresponding figure legends should be consulted for details of each particular illustration and analysis. The free, general-purpose NIH ImageJ software platform, its Fiji distribution package  and its TrakEM2 companion  provide excellent starting points for AT image analysis, as this platform is flexible and very suitable for the processing and analysis of high-dimensional AT images. Numerous commercial software solutions may better suit specific analysis scenarios, however, and analysis of the larger, high-resolution, high-content AT image datasets will necessarily tend toward requiring web- and cloud-based solutions to data storage, processing, and analysis challenges. Machine learning tools, including deep convolutional networks, are now revolutionizing all forms of volumetric image analysis, AT analysis included .
As a volumetric imaging modality, AT poses both challenges and opportunities for the computational segmentation of biologically meaningful three-dimensional objects, and for quantification of such objects. In the realm of circuit neuroscience, such image analyses have focused on detection of synapses, profiling of diverse synapse populations, and tracing of axons and dendrites. The high-dimensional molecular discrimination capacities realized in AT also pose exciting new opportunities to fathom the brain’s intricate molecular architectures, but with these will certainly come new challenges to data analysis.
Improved materials and methods for AT are rapidly increasing the power of AT and shrinking the technical difficulty that has so far limited applications. Ongoing materials engineering aims to improve the scope and sensitivity of AT molecular analysis: (1) resin chemistry to improve label access to embedded tissue proteins; (2) new protein labeling reagents, such as array-screened monoclonals, camelid antibodies, nanobodies, and recombinant immunoglobulin fragments to improve protein detection; (3) new organic and biologic fluors, fluorescent nanoparticles, cathodoluminescent tags, and DNA-barcoded antibody tags; and (4) new resin and probe chemistries to enable in situ mRNA hybridization with immunolabeling at AT resolution. As these new methods and others advance, technical and economic barriers to AT application should fall and enable the development of many new AT applications inside and outside of neuroscience. As AT methodologies become more routine and less expensive, application areas may eventually even grow to include clinical pathology.
The economic obstacles to large-scale AT should fall. With successes of ongoing AT process and tool engineering, commercialization advances, and continuation of the “Moore’s Law” deflation of computing costs, it can be anticipated that AT imaging of the microliter-scale volumes needed for local circuit connectomics and certain other tissue analysis challenges may eventually become routine. For large-scale, high-resolution digital microscopy, image acquisition times quickly become the rate-limiting step. (Several microliter-scale volumetric EM projects now under way envision image acquisition times on the order of months to years!) Here, AT offers substantial speed advantages in comparison to other comparable methods, because of AT’s high optical imaging efficiency. The engineering prototype illustrated in Fig. 8c acquires 1 aL, super-diffraction FM-AT voxels at an overall net rate well in excess of 10 million per second. The multibeam scanning electron microscope (mSEM) offers the prospect of acquiring EM-AT volume images at rates approaching one billion 0.001 aL EM-AT voxels per second.
Perhaps the most vibrant near-term growth in AT applications will build on the rapid advance of mRNA sequencing technologies. A flood of deep, single-cell transcriptomic data has ushered in the prospect of classifying neurons and other tissue cells into taxonomies comprising modest numbers of relatively discrete cell types, each complete with its own distinctive “parts list” of protein products predicted from gene expression patterns. At the simplest level, single-cell transcriptomic data and cell-type taxonomies will guide the selection of AT antibodies to more deeply and efficiently explore cell-type-specific molecular mechanisms in tissue architectural context. The AT superlatives of resolution, volume scalability, and molecular multiplexing are also likely to prove excellent fits to the challenges of cross-validating transcriptomic taxonomies to other dimensions of cell type differentiation (e.g., anatomy, physiology, proteomics, connectomics, or synaptomics). The prospects for AT cross-validation of cell-type and synapse-type taxonomies will likely grow even faster if resin chemistry developments enable some form of highly multiplexed RNA-FISH imaging compatible with existing immunofluorescence imaging capacities. In any case, detection of proteins predicted by mRNA transcript detection should strongly advance cross-validation of molecular taxonomies and offer new and fundamental insights into quantitative transcript–protein relationships. The strengths of AT imaging appear to match the needs of an approaching post-transcriptomic tissue science era very nicely indeed!
The author wishes to thank Allen Institute for Brain Science founders, Paul G. Allen and Jody Allen, for their vision and encouragement. This writing benefitted from helpful discussions over many years with colleagues including (but not limited to) Kristina Micheva and Forrest Collman, and Alex Baden, Ben Barres, Lynne Becker, Erik Bloss, Davi Bock, JoAnn Buchanan, Randal Burns, Brad Busse, Albert Cardona, Winfried Denk, Belvin Gong, Michael Hausser, Michael Hawrylycz, Allan Jones, Erik Jorgensen, Larry Kaplan, Christof Koch, Daphne Koller, Ed Lein, David Lenzi, Jeff Lichtman, Michael Linhoff, Liqun Luo, Daniel Madison, Gail Mandel, Steve McConoughey, Eugene Myers, William Mobley, Jon Mulholland, Andrew Olson, Nancy O’Rourke, Hanchuan Peng, Eric Perlman, Tom Reese, Estuardo Robles, Stephan Saalfeld, Guillermo Sapiro, Carla Shatz, William Seeley, James Sharp, Anish Simhal, Tara Spires-Jones, Nelson Spruston, Beth Stevens, Karel Svoboda, Bosiljka Tasic, Mark Terasaki, Jay Trautman, James Trimmer, Joshua Vogelstein, Gordon Wang, Shigeki Watanabe, Nicholas Weiler, Richard Weinberg, Stacie Weninger, Hongkui Zeng, and Joshua Zimmerberg. The author is also grateful to the authors and publishers who granted permissions to reproduce their published figures here and to Kristina Micheva for sharing unpublished data. Finally, the author gratefully acknowledges Dan Castelli, Sarah Davis, Leila Elabbady, Olga Gliko, Mats Karlsson, Michelle Naugle, Jenna Schardt, Martin Schroeder, Marie Scott, Rob Serafin, Sharmistaa Seshamani, Kristen Turner, Derric Williams, and the leadership of the Allen Institute’s Engineering departments for their creative contributions to the development of novel AT tools (including those illustrated in Fig. 8) at the Allen Institute.
The author is grateful for support from Allen Institute for Brain Science founders, Paul G. Allen and Jody Allen, and from the National Institutes of Health (grants R01NS092474 and R01MH104227).
The author read and approved the final manuscript after input from the editorial office.
The author has a founder’s equity interest in Aratome, LLC (Menlo Park, CA), an enterprise that markets AT materials and services.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Wacker I, Spomer W, Hofmann A, Thaler M, Hillmer S, Gengenbach U, Schröder RR. Hierarchical imaging: a new concept for targeted imaging of large volumes from cells to tissues. BMC Cell Biol. 2016;17:38.PubMedPubMed CentralView ArticleGoogle Scholar
- Wacker I, Spomer W, Hofmann A, Gengenbach U, Thaler M, Ness L, Brey P, Schroder RR. On the road to large volumes in LM and SEM: new tools for Array tomography. Microsc Microanal. 2015;21:539–40.View ArticleGoogle Scholar
- Markert SM, Bauer V, Muenz TS, Jones NG, Helmprobst F, Britz S, Sauer M, Rössler W, Engstler M, Stigloher C. 3D subcellular localization with superresolution array tomography on ultrathin sections of various species. Methods Cell Biol. 2017;140:21–47.PubMedView ArticleGoogle Scholar
- García-Fernández E, Koch G, Wagner RM, Fekete A, Stengel ST, Schneider J, Mielich-Süss B, Geibel S, Markert SM, Stigloher C, Lopez D. Membrane Microdomain Disassembly Inhibits MRSA Antibiotic Resistance. Cell. 2017;171:1354–1367.e20.PubMedPubMed CentralView ArticleGoogle Scholar
- Terasaki M, Shemesh T, Kasthuri N, Klemm RW, Schalek R, Hayworth KJ, Hand AR, Yankova M, Huber G, Lichtman JW, Rapoport T, Kozlov MM. Stacked endoplasmic reticulum sheets are connected by helicoidal membrane motifs. Cell. 2013;154:285–96.PubMedPubMed CentralView ArticleGoogle Scholar
- Wacker I, Schroeder RR. Array tomography. J Microsc. 2013;252:93–9.PubMedView ArticleGoogle Scholar
- Wacker I, Chockley P, Bartels C, Spomer W, Hofmann A, Gengenbach U, Singh S, Thaler M, Grabher C, Schröder RR. Array tomography: characterizing FAC-sorted populations of zebrafish immune cells by their 3D ultrastructure. J Microsc. 2015;259:105–13.PubMedPubMed CentralView ArticleGoogle Scholar
- Smith SM, Yarwood A, Fleck RA, Robinson C, Smith CJ. TatA complexes exhibit a marked change in organisation in response to expression of the TatBC complex. Biochem J. 2017;474:1495–508.PubMedPubMed CentralView ArticleGoogle Scholar
- Saatchi S, Azuma J, Wanchoo N, Smith SJ, Yock PG, Taylor CA, Tsao PS. Three-dimensional microstructural changes in murine abdominal aortic aneurysms quantified using immunofluorescent array tomography. J Histochem Cytochem. 2012;60:97–109.PubMedPubMed CentralView ArticleGoogle Scholar
- Peddie CJ, Collinson LM. Exploring the third dimension: volume electron microscopy comes of age. Micron. 2014;61:9–19.PubMedView ArticleGoogle Scholar
- Oga DK, Usumi SK, Shiki TU, Atanabe TW. Integrative method for three-dimensional imaging of the entire Golgi appa- ratus by combining thiamine pyrophosphatase cytochemistry and array to- mography using backscattered electron-mode scanning electron microscopy. Biomed Res. 2017;38:285–96.View ArticleGoogle Scholar
- Ochs M, Knudsen L, Hegermann J, Wrede C, Grothausmann R, Mühlfeld C. Using electron microscopes to look into the lung. Histochem Cell Biol. 2016;146:695–707.PubMedView ArticleGoogle Scholar
- Nanguneri S, Flottmann B, Horstmann H, Heilemann M, Kuner T. Three-dimensional, tomographic super-resolution fluorescence imaging of serially sectioned thick samples. PLoS One. 2012;7:1–8.View ArticleGoogle Scholar
- Micheva KD, Bruchez MP. The gain in brain: novel imaging techniques and multiplexed proteomic imaging of brain tissue ultrastructure. Curr Opin Neurobiol. 2012;22:94–100.PubMedView ArticleGoogle Scholar
- Prieto GA, Cotman CW. On the road towards the global analysis of human synapses. Neural Regen Res. 2017;12:1586–89Google Scholar
- Peddie CJ, Liv N, Hoogenboom JP, Collinson LM. Integrated light and scanning electron microscopy of GFP-expressing cells. Methods Cell Biol. 2014;124:363–9.PubMedView ArticleGoogle Scholar
- Micheva KD, Smith SJ. Array tomography: a new tool for imaging the molecular architecture and utrastructure of neural circuits. Neuron. 2007;55:25–36.PubMedPubMed CentralView ArticleGoogle Scholar
- Marc RE, Jones BW. Molecular phenotyping of retinal ganglion cells. J Neurosci. 2002;22:413–27.PubMedView ArticleGoogle Scholar
- Wahlby C, Erlandsson F, Bengtsson E, Zetterberg A. Sequential immunofluorescence staining and image analysis for detection of large numbers of antigens in individual cell nuclei. Cytometry. 2002;47:32–41.PubMedView ArticleGoogle Scholar
- Nanci A, Zalzal S, Smith CE. Routine use of backscattered electron imaging to visualize cytochemical and autoradiographic reactions in semi-thin plastic sections. J Histochem Cytochem. 1990;38:403–14.PubMedView ArticleGoogle Scholar
- Kasthuri N, Hayworth K, Lichtman JW, Erdman N, Ackerley C. New technique for ultra-thin serial brain section imaging using scanning electron microscopy. Microsc Microanal. 2007;13:21–3.View ArticleGoogle Scholar
- Punge A, Rizzoli SO, Jahn R, Wildanger JD, Meyer L, Schönle A, Kastrup L, Hell SW. 3D reconstruction of high-resolution STED microscope images. Microsc Res Tech. 2008;71:644–50.PubMedView ArticleGoogle Scholar
- De Rosier DJ, Klug A. Reconstruction of three dimensional structures from electron micrographs. Nature. 1968;217:130–4.PubMedView ArticleGoogle Scholar
- Hounsfield GN. Computed medical imaging. Med Phys. 1980;7:283–90.PubMedView ArticleGoogle Scholar
- Briggman KL, Bock DD. Volume electron microscopy for neuronal circuit reconstruction. Curr Opin Neurobiol. 2012;22:154–61.PubMedView ArticleGoogle Scholar
- Denk W, Briggman KL, Helmstaedter M. Structural neurobiology: missing link to a mechanistic understanding of neural computation. Nat Rev Neurosci. 2012;13:351–8.PubMedView ArticleGoogle Scholar
- Bloss EB, Cembrowski MS, Karsh B, Colonell J, Fetter RD, Spruston N. Structured dendritic inhibition supports branch-selective integration in CA1 pyramidal cells. Neuron. 2016;89:1016–30.PubMedView ArticleGoogle Scholar
- Rah J-C, Bas E, Colonell J, Mishchenko Y, Karsh B, Fetter RD, Myers EW, Chklovskii DB, Svoboda K, Harris TD, Isaac JTR. Thalamocortical input onto layer 5 pyramidal neurons measured using quantitative large-scale array tomography. Front Neural Circuits. 2013;7:177.PubMedPubMed CentralView ArticleGoogle Scholar
- Kubota Y, Sohn J, Hatada S, Schurr M, Straehle J, Gour A, Neujahr R, Miki T, Mikula S, Kawaguchi Y. A carbon nanotube tape for serial-section electron microscopy of brain ultrastructure. Nat Commun. 2018;9:1–3.View ArticleGoogle Scholar
- Oberti D, Kirschmann MA, Hanhloser R. Projection neuron circuits resolved using correlative array tomography. Front Neurosci. 2011;5:1–8.View ArticleGoogle Scholar
- Collman F, Buchanan J, Phend KD, Micheva KD, Weinberg RJ, Smith SJ. Mapping synapses by conjugate light-Electron Array tomography. J Neurosci. 2015;35:5792–807.PubMedPubMed CentralView ArticleGoogle Scholar
- Markert SM, Britz S, Proppert S, Lang M, Witvliet D, Mulcahy B, Sauer M, Zhen M, Bessereau J-L, Stigloher C. Filling the gap: adding super-resolution to array tomography for correlated ultrastructural and molecular identification of electrical synapses at the C. elegans connectome. Neurophotonics. 2016;3:041802.PubMedPubMed CentralView ArticleGoogle Scholar
- Wacker I, Schröder RR, Schroeder JA. Pathology goes 3D: exploring the potential of array tomography versus FIB nanotomography for a CADASIL sample. Ultrastruct Pathol. 2017;41:114–5.View ArticleGoogle Scholar
- Micheva KD, Wolman D, Mensh BD, Pax E, Buchanan J, Smith SJ, Bock DD. A large fraction of neocortical myelin ensheathes axons of local inhibitory neurons. elife. 2016;5:1–29.View ArticleGoogle Scholar
- Simhal AK, Aguerrebere C, Collman F, Vogelstein JT, Micheva KD, Weinberg RJ, Smith SJ, Sapiro G. Probabilistic fluorescence-based synapse detection. PLoS Comput Biol. 2016;13:1–24.Google Scholar
- Delpiano J, Pizarro L, Peddie CJ, Jones ML, Griffin LD, Collinson LM. Automated detection of fluorescent cells in in-resin fluorescence sections for integrated light and electron microscopy. J Microsc. 2018;271(1):109–19.PubMedPubMed CentralView ArticleGoogle Scholar
- Micheva KD, Busse B, Weiler NC, O’Rourke N, Smith SJ. Single-synapse analysis of a diverse synapse population: proteomic imaging methods and markers. Neuron. 2010;68:639–53.PubMedPubMed CentralView ArticleGoogle Scholar
- Müller-Reichert T, Verkade P. Correlative Light and Electron Microscopy II. Methods Cell Biol. 2014;124:xvii–xviii.PubMedView ArticleGoogle Scholar
- D’Andrea MR. Chapter 4 – Addressing Technical Concerns. Bursting Neurons and Fading Memories; 2015. p. 33–41.View ArticleGoogle Scholar
- Newman GR, Hobot JA. Resins for combined light and electron microscopy: a half century of development. Histochem J. 1999;31:495–505.PubMedView ArticleGoogle Scholar
- Schalek R, Wilson A, Lichtman JW, Josh M, Kasthuri N, Berger D, Seung HS, Anger P, Hayworth K, Aderhold D. ATUM-based SEM for high-speed large-volume biological reconstructions. Microsc Microanal. 2012;18:572–3.View ArticleGoogle Scholar
- Joesch M, Mankus D, Yamagata M, Shahbazi A, Schalek R, Suissa-Peleg A, Meister M, Lichtman JW, Scheirer WJ, Sanes JR. Reconstruction of genetically identified neurons imaged by serial-section electron microscopy. elife. 2016;5:e15015.PubMedPubMed CentralView ArticleGoogle Scholar
- Glenn DR, Zhang H, Kasthuri N, Schalek R, Lo PK, Trifonov AS, Park H, Lichtman JW, Walsworth RL. Correlative light and electron microscopy using cathodoluminescence from nanoparticles with distinguishable colours. Sci Rep. 2012;2:865.PubMedPubMed CentralView ArticleGoogle Scholar
- Watanabe S, Punge A, Hollopeter G, Willig KI, Hobson RJ, Davis MW, Hell SW, Jorgensen EM. Protein localization in electron micrographs using fluorescence nanoscopy. Nat Methods. 2011;8:80–4.PubMedView ArticleGoogle Scholar
- Müller-Reichert T, Verkade P. Correlative Light and Elelctron Microscopy. Cambridge: Academic Press; 2012.Google Scholar
- McDonald K, Schwarz H, Müller-Reichert T, Webb R, Buser C, Morphew M. “Tips and tricks” for high-pressure freezing of model systems. Methods Cell Biol. 2010;96:671–93.PubMedView ArticleGoogle Scholar
- Webster P, Schwarz H, Griffiths G. Preparation of cells and tissues for Immuno EM. Methods Cell Biol. 2008;88:45–58.PubMedView ArticleGoogle Scholar
- Watanabe S, Jorgensen EM. Visualizing Proteins in Electron Micrographs at Nanometer Resolution. In: Müller-Reichert T, Verkade P, editors. Correlative Light Electron Microscopy. Cambridge: Academic Press; 2012. p. 283–306.Google Scholar
- Hayworth KJ, Morgan JL, Schalek R, Berger DR, Hildebrand DGC, Lichtman JW. Imaging ATUM ultrathin section libraries with WaferMapper: a multi-scale approach to EM reconstruction of neural circuits. Front Neural Circuits. 2014;8:68.PubMedPubMed CentralView ArticleGoogle Scholar
- RMC_Boeckler. ATUMtome: Automated Tape Collecting Ultramicrotome. https://www.rmcboeckeler.com/atumtome/. Accessed 30 July 2018.
- Koike T, Kataoka Y, Maeda M, Hasebe Y, Yamaguchi Y, Suga M, Saito A, Yamada H. A device for ribbon collection for Array tomography with scanning Electron microscopy. Acta Histochem Cytochem. 2017;50:135–40.PubMedPubMed CentralView ArticleGoogle Scholar
- Spomer W, Hofmann A, Wacker I, Ness L, Brey P, Schroder RR, Gengenbach U. Advanced substrate holder and multi-axis manipulation tool for ultramicrotomy. Microsc Microanal. 2015;21:1277–8.View ArticleGoogle Scholar
- RMC_Boeckler. ASH-100 Advanced Substrate Holder. https://www.rmcboeckeler.com/ash-100/. Accessed 30 July 2018.
- Templier T, Hanhloser R. Automated dense collection of ultrathin sections directly onto silicon wafers. Neurosci Meet Planner San Diego, CA Soc Neurosci Online http://www.abstractsonline.com/pp8/index.html#!/4071/presentation/25775. 2016. p. Program No. 186.08. Accessed 30 July 2018.
- Kasthuri N, Hayworth KJ, Berger DR, Schalek RL, Conchello JA, Knowles-Barley S, Lee D, Vázquez-Reina A, Kaynig V, Jones TR, Roberts M, Morgan JL, Tapia JC, Seung HS, Roncal WG, Vogelstein JT, et al. Saturated reconstruction of a volume of neocortex. Cell. 2015;162:648–61.PubMedView ArticleGoogle Scholar
- Linhoff MW, Garg SK, Mandel G. A high-resolution imaging approach to investigate chromatin architecture in complex tissues. Cell. 2015;163:246–55.PubMedPubMed CentralView ArticleGoogle Scholar
- Maxwell DJ, Christie WM, Somogyi P. Synaptic connections of GABA-containing boutons in the lateral cervical nucleus of the cat: an ultrastructural study employing pre- and post-embedding immunocytochemical methods. Neuroscience. 1989;33:169–84.PubMedView ArticleGoogle Scholar
- Griffiths G, Burke B, Lucocq J. Fine Structure Immunocytochemistry. Berlin: Springer-Verlag; 1993.Google Scholar
- Micheva KD, Phend KD. Conjugate immunofluorescence — SEM Array Tomography for Studying Mammalian Synapses and Axons. In: Hanssen E, editor. Cellular imaging. Cham: Springer International Publishing AG; 2018. p. 149–82.Google Scholar
- Immunofluorescent array tomography webinar. http://www.abcam.com/webinars/immunofluorescent-array-tomography-webinar. Accessed 30 July 2018.
- Neuromab utility for array tomography. http://neuromab.ucdavis.edu/arrayTomography.cfm. Accessed 30 July 2018.
- Simhal AK, Gong B, Trimmer JS, Weinberg RJ, Smith SJ, Sapiro G, Micheva KD. A Computational Synaptic Antibody Characterization and Screening Framework for Array Tomography. bioRxiv. 2018:258756.Google Scholar
- Sigal YM, Speer CM, Babcock HP, Zhuang X. Mapping synaptic input fields of neurons with super-resolution imaging. Cell. 2015;163:493–505.PubMedPubMed CentralView ArticleGoogle Scholar
- ZEISS ZEN Correlative Array Tomography. https://www.zeiss.com/microscopy/int/products/correlative-microscopy/zen-correlative-array-tomography.html. Accessed 30 July 2018.
- Carl_Zeiss_Microscopy. Three-Dimensional Scanning Electron Microscopy for Biology - Bitesize Bio. http://bitesizebio.com/30796/three-dimensional-scanning-electron-microscopy-for-biology/. Accessed 30 July 2018.
- Yoo HW, Van Royen ME, Van Cappellen WA, Houtsmuller AB, Verhaegen M, Schitter G. Automated spherical aberration correction in scanning confocal microscopy. Rev Sci Instrum. 2014;85:123706.PubMedView ArticleGoogle Scholar
- Wang GX, Smith SJ. Sub-diffraction limit localization of proteins in volumetric space using Bayesian restoration of fluorescence images from ultrathin specimens. PLoS Comput Biol. 2012;8:e1002671.PubMedPubMed CentralView ArticleGoogle Scholar
- McGorty R, Schnitzbauer J, Zhang W, Huang B. Correction of depth-dependent aberrations in 3D single molecule localization and super-resolution microscopy. Opt Lett. 2014;39:275–8.PubMedPubMed CentralView ArticleGoogle Scholar
- Hong S, Wilton DK, Stevens B, Richardson DS. Structured illumination microscopy for the investigation of synaptic structure and function. Methods Mol Biol. 2017;1538:155–67.PubMedPubMed CentralView ArticleGoogle Scholar
- Ullal AV, Weissleder R. Single cell protein analysis. Methods Mol Biol. 2015;1346:47–54.PubMedView ArticleGoogle Scholar
- Lubeck E, Coskun AF, Zhiyentayev T, Ahmad M, Cai L. Correspondence: single-cell in situ RNA profiling by sequential hybridization. Nat Methods. 2014;11:360–1.PubMedPubMed CentralView ArticleGoogle Scholar
- Model M. Intensity calibration and flat-field correction for fluorescence microscopes. Curr Protoc Cytom. 2014;68:1–10.Google Scholar
- Thévenaz P, Ruttimann UE, Unser M. A pyramid approach to subpixel registration based on intensity. IEEE Trans Image Process. 1998;7:27–41.PubMedView ArticleGoogle Scholar
- Busse B, Miura K. MultiStackRegistration. 2013. https://github.com/miura/MultiStackRegistration. Accessed 30 July 2018.
- Saalfeld S, Fetter R, Cardona A, Tomancak P. Elastic volume reconstruction from series of ultra-thin microscopy sections. Nat Methods. 2012;9:717–20.PubMedView ArticleGoogle Scholar
- Ounkomol C, Seshamani S, Maleckar MM, Collman F, Johnson G. Label-free prediction of three-dimensional fluorescence images from transmitted light microscopy. bioRxiv. 2018.Google Scholar
- Vogelstein JT, Mensh B, Häusser M, Spruston N, Evans AC, Kording K, Amunts K, Ebell C, Muller J, Telefont M, Hill S, Koushika SP, Calì C, Valdés-Sosa PA, Littlewood PB, Koch C, et al. To the cloud! A grassroots proposal to accelerate brain science discovery. Neuron. 2016;92:622–7.PubMed CentralView ArticleGoogle Scholar
- Burns R, Vogelstein JT, Szalay ASS. From cosmos to connectomes: the evolution of data-intensive science. Neuron. 2014;83:1249–52.PubMedView ArticleGoogle Scholar
- Burns R, Roncal WG, Kleissas D, Lillaney K, Manavalan P, Perlman E, Berger DR, Bock DD, Chung K, Grosenick L, Kasthuri N, Weiler NC, Deisseroth K, Kazhdan M, Lichtman J, Reid RC, et al. The Open Connectome Project Data Cluster: Scalable Analysis and Vision for High-Throughput Neuroscience. Sci Stat Database Manag. 2013.Google Scholar
- Koffie RM, Meyer-Luehmann M, Hashimoto T, Adams KW, Mielke ML, Garcia-Alloza M, Micheva KD, Smith SJ, Kim ML, Lee VM, Hyman BT, Spires-Jones TL. Oligomeric amyloid beta associates with postsynaptic densities and correlates with excitatory synapse loss near senile plaques. Proc Natl Acad Sci U S A. 2009;106:4012–7.PubMedPubMed CentralView ArticleGoogle Scholar
- Mishchenko Y. On optical detection of densely labeled synapses in neuropil and mapping connectivity with combinatorially multiplexed fluorescent synaptic markers. PLoS One. 2010;5:e8853.PubMedPubMed CentralView ArticleGoogle Scholar
- Li L, Tasic B, Micheva KD, Ivanov VM, Spletter ML, Smith SJ, Luo L. Visualizing the distribution of synapses from individual neurons in the mouse brain. PLoS One. 2010;5:e11503.PubMedPubMed CentralView ArticleGoogle Scholar
- Kopeikina KJ, Carlson GA, Pitstick R, Ludvigson AE, Peters A, Luebke JI, Koffie RM, Frosch MP, Hyman BT, Spires-Jones TL. Tau accumulation causes mitochondrial distribution deficits in neurons in a mouse model of tauopathy and in human Alzheimer’s disease brain. Am J Pathol. 2011;179:2071–82.PubMedPubMed CentralView ArticleGoogle Scholar
- Allen NJ, Bennett ML, Foo LC, Wang GX, Chakraborty C, Smith SJ, Barres BA. Astrocyte glypicans 4 and 6 promote formation of excitatory synapses via GluA1 AMPA receptors. Nature. 2012;486:410–4.PubMedPubMed CentralView ArticleGoogle Scholar
- Busse B, Smith SJ. Automated analysis of a diverse synapse population. PLoS Comput Biol. 2013;9:e1002976.PubMedPubMed CentralView ArticleGoogle Scholar
- Rasakham K, Schmidt HD, Kay K, Huizenga MN, Calcagno N, Christopher Pierce R, Spires-Jones TL, Sadri-Vakili G. Synapse density and dendritic complexity are reduced in the prefrontal cortex following seven days of forced abstinence from cocaine self-administration. PLoS One. 2014;9:1–6.View ArticleGoogle Scholar
- Neuman KM, Molina-Campos E, Musial TF, Price AL, Oh KJ, Wolke ML, Buss EW, Scheff SW, Mufson EJ, Nicholson DA. Evidence for Alzheimer’s disease-linked synapse loss and compensation in mouse and human hippocampal CA1 pyramidal neurons. Brain Struct Funct. 2015;220:3143–65.PubMedView ArticleGoogle Scholar
- Rah J, Feng L, Druckmann S, Lee H, Kim J. From a meso- to micro-scale connectome: array tomography and mGRASP. Front Neuroanat. 2015;9:1–12.View ArticleGoogle Scholar
- Nosheny RL, Belichenko PV, Busse B, Weissmiller M, Dang V, Das D, Fahimi A, Salehi A, Smith SJ, Mobley WC. Increased cortical synaptic activation of TrkB and downstream signaling markers in a mouse model of Down syndrome. Neurobiol Dis. 2015;77:173–90.PubMedPubMed CentralView ArticleGoogle Scholar
- Mierau SB, Patrizi A, Hensch TK, Fagiolini M. Cell-specific regulation of N-methyl-D-aspartate receptor maturation by Mecp2 in cortical circuits. Biol Psychiatry. 2016;79:746–54.PubMedView ArticleGoogle Scholar
- Jackson RJ, Rudinskiy N, Herrmann AG, Croft S, Kim JSM, Petrova V, Ramos-Rodriguez JJ, Pitstick R, Wegmann S, Garcia-Alloza M, Carlson GA, Hyman BT, Spires-Jones TL. Human tau increases amyloid β plaque size but not amyloid β-mediated synapse loss in a novel mouse model of Alzheimer’s disease. Eur J Neurosci. 2016;44:3056–66.PubMedPubMed CentralView ArticleGoogle Scholar
- Morgan JL, Berger DRR, Wetzel AWW, Lichtman JW. The fuzzy logic of network connectivity in mouse visual thalamus. Cell. 2016;165:192–206.PubMedPubMed CentralView ArticleGoogle Scholar
- Wang GX, Smith SJ, Mourrain P. Sub-synaptic, multiplexed analysis of proteins reveals fragile X related protein 2 is mislocalized in Fmr1 KO synapses. elife. 2016;5:1–20.Google Scholar
- Pickett EK, Henstridge CM, Allison E, Pitstick R, Pooler A, Wegmann S, Carlson G, Hyman BT, Spires-Jones TL. Spread of tau down neural circuits precedes synapse and neuronal loss in the rTgTauEC mouse model of early Alzheimer’s disease. Synapse. 2017;71:1–8.View ArticleGoogle Scholar
- Henstridge CM, Sideris DI, Carroll E, Rotariu S, Salomonsson S, Tzioras M, McKenzie CA, Smith C, von Arnim CAF, Ludolph AC, Lulé D, Leighton D, Warner J, Cleary E, Newton J, Swingler R, et al. Synapse loss in the prefrontal cortex is associated with cognitive decline in amyotrophic lateral sclerosis. Acta Neuropathol. 2017;135:1–14.Google Scholar
- Bloss EB, Cembrowski MS, Karsh B, Colonell J, Fetter RD, Spruston N. Single excitatory axons form clustered synapses onto CA1 pyramidal cell dendrites. Nat Neurosci. 2018;21(3):353–63.PubMedView ArticleGoogle Scholar
- Valenzuela RA, Micheva KD, Kiraly M, Li D, Madison DV. Array tomography of physiologically-characterized CNS synapses. J Neurosci Methods. 2016;268:43–52.PubMedPubMed CentralView ArticleGoogle Scholar
- Datwani A, McConnell MJ, Kanold PO, Micheva KD, Busse B, Shamloo M, Smith SJ, Shatz CJ. Classical MHCI molecules regulate Retinogeniculate refinement and limit ocular dominance plasticity. Neuron. 2009;64:463–70.PubMedPubMed CentralView ArticleGoogle Scholar
- Schindelin J, Arganda-Carreras I, Frise E, Kaynig V, Longair M, Pietzsch T, Preibisch S, Rueden C, Saalfeld S, Schmid B, Tinevez JY, White DJ, Hartenstein V, Eliceiri K, Tomancak P, Cardona A. Fiji: an open-source platform for biological-image analysis. Nat Methods. 2012;9:676–82.PubMedView ArticlePubMed CentralGoogle Scholar
- Cardona A, Saalfeld S, Schindelin J, Arganda-Carreras I, Preibisch S, Longair M, Tomancak P, Hartenstein V, Douglas RJ. TrakEM2 software for neural circuit reconstruction. PLoS One. 2012;7:e28011.Google Scholar
- Stevens B, Allen NJ, Vazquez LE, Howell GR, Christopherson KS, Nouri N, Micheva KD, Mehalow AK, Huberman AD, Stafford B, Sher A, Litke A, Lambris JD, Smith SJ, John SWM, Barres BA. The classical complement Cascade mediates CNS synapse elimination. Cell. 2007;131:1164–78.PubMedView ArticleGoogle Scholar
- Soiza-Reilly M, Commons KG. Unraveling the architecture of the dorsal raphe synaptic neuropil using high-resolution neuroanatomy. Front Neural Circuits. 2014;8:1–7.View ArticleGoogle Scholar
- Madisen L, Zwingman TA, Sunkin SM, Oh SW, Zariwala HA, Gu H, Ng LL, Palmiter RD, Hawrylycz MJ, Jones AR, Lein ES, Zeng H. A robust and high-throughput Cre reporting and characterization system for the whole mouse brain. Nat Neurosci. 2010;13:133–40.PubMedView ArticleGoogle Scholar
- Ma Y, Hof PR, Grant SC, Blackband SJ, Bennett R, Slatest L, Mcguigan MD, Benveniste H. A three-dimensional digital atlas database of the adult C57BL/6J mouse brain by magnetic resonance microscopy. Neuroscience. 2005;135:1203–15.PubMedView ArticleGoogle Scholar